SaveOpen cresol12.dec 12 [bereit fr 12/1]
BatGosub std [SP=2]
   Inert Argon
   OHq Wasser
   Reakt m-Cresol
   Scav O2 1
   Reakt3 NC 0
   SollSumflow 1300
   SollFlowZelle 0 9999
   WartDauer 300
   SollcOHq 2000
   SollcReakt 0
   SollcScav 0
   SollcReakt3 0
   SollSweeps 50
   SollblFrequ 0.25
   SollBlEnergie 2.4
   SolltEnd 4 [4096*0.977 ms]
   SollTzelle 300
   SollpZelle 133
   SollTrohr 299
   BatReturn  [SP=1]
Scav NO 2e-3
ZeigeAntoine [
  8.610  1948  -24.1    Wasser(0-19 Grad),
  8.150  2124  -59.0    m-cresol(C7H8O, St ]
ZeigeCalib [
        name   ad   da   io  offset   gain linear.    Bemerkung
       traum   16   -1   -1  273.50 120.400 0.0000  RTD(0!!!..600ø)
  tzellpt100   15   -1   -1  174.80  39.800 0.0000  RTD(-100..100ø)
        tbad   13   -1   -1  272.00 125.000 0.0000  K 20 -Thermostat
        tohq   17   -1   -1  174.80  39.600 0.0000  RTD(-100..100ø)
      treakt   18   -1   -1  174.70  39.600 0.0000  RTD(-100..100ø)
  pzelle1000    5   -1   -1    0.00 132.000 0.0000  MKS(0..1000mbar)
    pzelle10   -7   -1   -1   -0.30   1.381 0.0000  MKS(0..10torr)
  psetze1000   -1    4 1015    0.00 -266.600 0.0000  Setze Pzelle1000
    psetze10   -1    4  -15   -0.30  -2.000 0.0000  Setze Pzelle10
        psat    6   -1   -1    0.00 135.000 0.0000  MKS(0..1000mbar)
       flow1    0    0   -1   94.03 364.000 10.3230  Bo822527_2  2000Ar
       flow2    1    6    1    1.85  10.240 0.0950   Bo822530_5  50Ar
       flow3    2    8    0    0.12   1.186 -0.0314  A_192_84_1568 5Ar
       flow4    3   10    2   -0.10   0.791 -0.0097  A_192_88_0022 5Ar f
       flow5    4    2   -1    0.08  46.910 -0.6557  A_192_82_4885 200Ar
     blitzn2   -1   -1    4    0.00   0.000 0.0000  Magnetventil
   bltrigger   -1   -1   12    0.00   0.000 0.0000  PC30 an Zndspule ]
Scav 15 .01
Scav o2 2e-3
Scav O2 2e-3
Scav NO 2e-3
SollcScav 3
EinstKonz 
SollcScav 1.5
SollcScav 1
EinstKonz 
DosBefehl :t
Ende 
SaveOpen cresol12.dec 12 [bereit fr 12/1]
BatGosub std [SP=2]
   Inert Argon
   OHq Wasser
   Reakt m-Cresol
   Scav NO 2E-3
   Reakt3 NC 0
   SollSumflow 1300
   SollFlowZelle 0 9999
   WartDauer 300
   SollcOHq 2000
   SollcReakt 0
   SollcScav 0
   SollcReakt3 0
   SollSweeps 50
   SollblFrequ 0.25
   SollBlEnergie 2.4
   SolltEnd 4 [4096*0.977 ms]
   SollTzelle 300
   SollpZelle 133
   SollTrohr 299
   BatReturn  [SP=1]
ZeigeCalib [
        name   ad   da   io  offset   gain linear.    Bemerkung
       traum   16   -1   -1  273.50 120.400 0.0000  RTD(0!!!..600ø)
  tzellpt100   15   -1   -1  174.80  39.800 0.0000  RTD(-100..100ø)
        tbad   13   -1   -1  272.00 125.000 0.0000  K 20 -Thermostat
        tohq   17   -1   -1  174.80  39.600 0.0000  RTD(-100..100ø)
      treakt   18   -1   -1  174.70  39.600 0.0000  RTD(-100..100ø)
  pzelle1000    5   -1   -1    0.00 132.000 0.0000  MKS(0..1000mbar)
    pzelle10   -7   -1   -1   -0.30   1.381 0.0000  MKS(0..10torr)
  psetze1000   -1    4 1015    0.00 -266.600 0.0000  Setze Pzelle1000
    psetze10   -1    4  -15   -0.30  -2.000 0.0000  Setze Pzelle10
        psat    6   -1   -1    0.00 135.000 0.0000  MKS(0..1000mbar)
       flow1    0    0   -1   94.03 364.000 10.3230  Bo822527_2  2000Ar
       flow2    1    6    1    1.85  10.240 0.0950   Bo822530_5  50Ar
       flow3    2    8    0    0.12   1.186 -0.0314  A_192_84_1568 5Ar
       flow4    3   10    2   -0.10   0.791 -0.0097  A_192_88_0022 5Ar f
       flow5    4    2   -1    0.08  46.910 -0.6557  A_192_82_4885 200Ar
     blitzn2   -1   -1    4    0.00   0.000 0.0000  Magnetventil
   bltrigger   -1   -1   12    0.00   0.000 0.0000  PC30 an Zndspule ]
ZeigeAntoine [
  8.610  1948  -24.1    Wasser(0-19 Grad),
  8.150  2124  -59.0    m-cresol(C7H8O, St ]
ZeigeZeit  [2.9. 14:55]
ZeigeZeit  [2.9. 14:55]
Messung  [Abbruch von Hand nach 13(50) Sweeps]
SollcScav 15
EinstKonz 
Messung  [Abbruch von Hand nach 15(50) Sweeps]
SollcReakt 3
EinstKonz 
Messung  [Abbruch von Hand nach 13(50) Sweeps]
DosBefehl :t w.
SollcReakt 0
SollcScav 0
EinstKonz 
[
SollcReakt 0
WartDauer 300
SollSweeps 100  SolltEnd 5
     SollcScav 15 BatGosub m
     SollcScav  1 BatGosub m
     SollcScav 15 BatGosub m
     SollcScav   1  BatGosub m
     SollcScav   2  BatGosub m
     SollcScav   3  BatGosub m
     SollcScav   5  BatGosub m
     SollcScav   8  BatGosub m
     SollcScav  10  BatGosub m
     SollcScav  12  BatGosub m
     SollcScav  15  BatGosub m
WartDauer 1200  SolltEnd 3
SollcReakt 3
     SollcScav   0  BatGosub m
WartDauer 300
     SollcScav   1  BatGosub m
     SollcScav   2  BatGosub m
     SollcScav   3  BatGosub m
     SollcScav   5  BatGosub m
     SollcScav   8  BatGosub m
     SollcScav  10  BatGosub m
     SollcScav  12  BatGosub m
     SollcScav  15  BatGosub m
     SollcScav  15  BatGosub m
     SollcScav  12  BatGosub m
     SollcScav  10  BatGosub m
     SollcScav   8  BatGosub m
     SollcScav   5  BatGosub m
     SollcScav   3  BatGosub m
     SollcScav   2  BatGosub m
     SollcScav   1  BatGosub m
     SollcScav   0  BatGosub m
WartDauer 1200  SolltEnd 3
SollcReakt 6
     SollcScav   0  BatGosub m
WartDauer 300
     SollcScav   1  BatGosub m
     SollcScav   2  BatGosub m
     SollcScav   3  BatGosub m
     SollcScav   5  BatGosub m
     SollcScav   8  BatGosub m
     SollcScav  10  BatGosub m
     SollcScav  12  BatGosub m
     SollcScav  15  BatGosub m
     SollcScav  15  BatGosub m
     SollcScav  12  BatGosub m
     SollcScav  10  BatGosub m
     SollcScav   8  BatGosub m
     SollcScav   5  BatGosub m
     SollcScav   3  BatGosub m
     SollcScav   2  BatGosub m
     SollcScav   1  BatGosub m
     SollcScav   0  BatGosub m
WartDauer 1200  SolltEnd 5
SollcReakt    0    BatGosub m
SollcReakt    0    Batgosub m
]
BatGosub w. [SP=2]
   SollcReakt 0
   WartDauer 300
   SollSweeps 100
   SolltEnd 5 [4096*1.221 ms]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 15:41, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
1 Traum     :  0.196     297.05 K
1 Tzelle    :  3.790     325.62 K     300 K
1 Tbad      :  0.002     272.19 K
1 TOHq      :  2.926     290.69 K    20.0813 mbar
1 TReakt    :  3.438     310.84 K     0.5223 mbar
1 pSat      :  7.472    1008.74 mbar
1 pZelle1000:  0.964     127.22 mbar   133 mbar
1 pZelle10  :  0.000       0.00 mbar
1 Flow1     :  2.938    1252.58 sccm
1 Flow2     :  4.079      45.20 sccm     Wasser:     1995.39     2000.00
1 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
1 Flow4     :  4.784       3.46 sccm         NO:       15.03       15.00
1 Flow5     : -0.003       0.00 sccm
1       sumflow   :     1302.15 sccm  1300.00 sccm
1       dosierFlow:     1302.15 sccm  9999.00 sccm
1       flowZelle :       10369  ccm        0  ccm
1 4096*Dwell= 5.001 s    100 Sweeps
1 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/1]
      BatReturn  [SP=2]
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 15:57, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
2 Traum     :  0.195     297.02 K
2 Tzelle    :  3.790     325.62 K     300 K
2 Tbad      :  0.002     272.22 K
2 TOHq      :  2.926     290.67 K    20.0616 mbar
2 TReakt    :  3.438     310.84 K     0.5223 mbar
2 pSat      :  7.473    1008.91 mbar
2 pZelle1000:  0.964     127.20 mbar   133 mbar
2 pZelle10  :  0.000       0.00 mbar
2 Flow1     :  2.938    1252.56 sccm
2 Flow2     :  4.079      45.20 sccm     Wasser:     1997.62     2000.00
2 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
2 Flow4     :  0.449       0.25 sccm         NO:        1.09        1.00
2 Flow5     : -0.003       0.00 sccm
2       sumflow   :     1298.92 sccm  1300.00 sccm
2       dosierFlow:     1298.92 sccm  9999.00 sccm
2       flowZelle :       10344  ccm        0  ccm
2 4096*Dwell= 5.001 s    100 Sweeps
2 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/2]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 16:13, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
3 Traum     :  0.196     297.06 K
3 Tzelle    :  3.789     325.62 K     300 K
3 Tbad      :  0.001     272.19 K
3 TOHq      :  2.926     290.65 K    20.0410 mbar
3 TReakt    :  3.438     310.84 K     0.5225 mbar
3 pSat      :  7.471    1008.57 mbar
3 pZelle1000:  0.964     127.21 mbar   133 mbar
3 pZelle10  :  0.000       0.00 mbar
3 Flow1     :  2.937    1252.01 sccm
3 Flow2     :  4.120      45.65 sccm     Wasser:     2011.75     2000.00
3 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
3 Flow4     :  4.784       3.46 sccm         NO:       15.03       15.00
3 Flow5     : -0.003       0.00 sccm
3       sumflow   :     1302.05 sccm  1300.00 sccm
3       dosierFlow:     1302.05 sccm  9999.00 sccm
3       flowZelle :       10368  ccm        0  ccm
3 4096*Dwell= 5.001 s    100 Sweeps
3 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/3]
      BatReturn  [SP=2]
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 16:28, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
4 Traum     :  0.195     297.02 K
4 Tzelle    :  3.789     325.60 K     300 K
4 Tbad      :  0.002     272.23 K
4 TOHq      :  2.925     290.64 K    20.0205 mbar
4 TReakt    :  3.438     310.84 K     0.5224 mbar
4 pSat      :  7.470    1008.39 mbar
4 pZelle1000:  0.964     127.22 mbar   133 mbar
4 pZelle10  :  0.000       0.00 mbar
4 Flow1     :  2.937    1252.15 sccm
4 Flow2     :  4.080      45.21 sccm     Wasser:     1995.89     2000.00
4 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
4 Flow4     :  0.449       0.25 sccm         NO:        1.09        1.00
4 Flow5     : -0.003       0.00 sccm
4       sumflow   :     1298.52 sccm  1300.00 sccm
4       dosierFlow:     1298.52 sccm  9999.00 sccm
4       flowZelle :       10340  ccm        0  ccm
4 4096*Dwell= 5.001 s    100 Sweeps
4 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/4]
      BatReturn  [SP=2]
   SollcScav 2
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 16:44, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
5 Traum     :  0.195     297.02 K
5 Tzelle    :  3.789     325.60 K     300 K
5 Tbad      :  0.002     272.21 K
5 TOHq      :  2.926     290.65 K    20.0382 mbar
5 TReakt    :  3.438     310.85 K     0.5228 mbar
5 pSat      :  7.470    1008.43 mbar
5 pZelle1000:  0.964     127.18 mbar   133 mbar
5 pZelle10  :  0.000       0.00 mbar
5 Flow1     :  2.937    1252.12 sccm
5 Flow2     :  4.079      45.20 sccm     Wasser:     1996.39     2000.00
5 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
5 Flow4     :  0.745       0.48 sccm         NO:        2.09        2.00
5 Flow5     : -0.002       0.00 sccm
5       sumflow   :     1298.72 sccm  1300.00 sccm
5       dosierFlow:     1298.72 sccm  9999.00 sccm
5       flowZelle :       10344  ccm        0  ccm
5 4096*Dwell= 5.001 s    100 Sweeps
5 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/5]
      BatReturn  [SP=2]
   SollcScav 3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 17:00, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
6 Traum     :  0.196     297.05 K
6 Tzelle    :  3.789     325.59 K     300 K
6 Tbad      :  0.001     272.16 K
6 TOHq      :  2.926     290.65 K    20.0415 mbar
6 TReakt    :  3.438     310.85 K     0.5229 mbar
6 pSat      :  7.468    1008.23 mbar
6 pZelle1000:  0.964     127.20 mbar   133 mbar
6 pZelle10  :  0.000       0.00 mbar
6 Flow1     :  2.937    1252.15 sccm
6 Flow2     :  4.080      45.21 sccm     Wasser:     1997.48     2000.00
6 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
6 Flow4     :  0.973       0.66 sccm         NO:        2.86        3.00
6 Flow5     : -0.002       0.00 sccm
6       sumflow   :     1298.93 sccm  1300.00 sccm
6       dosierFlow:     1298.93 sccm  9999.00 sccm
6       flowZelle :       10344  ccm        0  ccm
6 4096*Dwell= 5.001 s    100 Sweeps
6 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/6]
      BatReturn  [SP=2]
   SollcScav 5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 17:15, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
7 Traum     :  0.195     297.02 K
7 Tzelle    :  3.789     325.60 K     300 K
7 Tbad      :  0.002     272.19 K
7 TOHq      :  2.926     290.65 K    20.0391 mbar
7 TReakt    :  3.438     310.85 K     0.5227 mbar
7 pSat      :  7.466    1007.97 mbar
7 pZelle1000:  0.965     127.42 mbar   133 mbar
7 pZelle10  :  0.000       0.00 mbar
7 Flow1     :  2.939    1252.95 sccm
7 Flow2     :  4.080      45.21 sccm     Wasser:     1999.54     2000.00
7 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
7 Flow4     :  1.578       1.12 sccm         NO:        4.88        5.00
7 Flow5     : -0.002       0.00 sccm
7       sumflow   :     1300.20 sccm  1300.00 sccm
7       dosierFlow:     1300.20 sccm  9999.00 sccm
7       flowZelle :       10337  ccm        0  ccm
7 4096*Dwell= 5.001 s    100 Sweeps
7 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/7]
      BatReturn  [SP=2]
   SollcScav 8
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 17:31, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
8 Traum     :  0.195     297.03 K
8 Tzelle    :  3.789     325.60 K     300 K
8 Tbad      :  0.001     272.18 K
8 TOHq      :  2.925     290.63 K    20.0138 mbar
8 TReakt    :  3.438     310.84 K     0.5226 mbar
8 pSat      :  7.464    1007.69 mbar
8 pZelle1000:  0.964     127.27 mbar   133 mbar
8 pZelle10  :  0.000       0.00 mbar
8 Flow1     :  2.940    1253.39 sccm
8 Flow2     :  4.121      45.66 sccm     Wasser:     2012.32     2000.00
8 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
8 Flow4     :  2.535       1.84 sccm         NO:        8.00        8.00
8 Flow5     : -0.002       0.00 sccm
8       sumflow   :     1301.81 sccm  1300.00 sccm
8       dosierFlow:     1301.81 sccm  9999.00 sccm
8       flowZelle :       10362  ccm        0  ccm
8 4096*Dwell= 5.001 s    100 Sweeps
8 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/8]
      BatReturn  [SP=2]
   SollcScav 10
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 17:47, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
9 Traum     :  0.195     297.02 K
9 Tzelle    :  3.789     325.60 K     300 K
9 Tbad      :  0.002     272.19 K
9 TOHq      :  2.925     290.63 K    20.0080 mbar
9 TReakt    :  3.438     310.85 K     0.5229 mbar
9 pSat      :  7.465    1007.71 mbar
9 pZelle1000:  0.964     127.21 mbar   133 mbar
9 pZelle10  :  0.000       0.00 mbar
9 Flow1     :  2.939    1252.84 sccm
9 Flow2     :  4.121      45.66 sccm     Wasser:     2010.97     2000.00
9 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
9 Flow4     :  3.192       2.32 sccm         NO:       10.10       10.00
9 Flow5     : -0.003       0.00 sccm
9       sumflow   :     1301.74 sccm  1300.00 sccm
9       dosierFlow:     1301.74 sccm  9999.00 sccm
9       flowZelle :       10366  ccm        0  ccm
9 4096*Dwell= 5.001 s    100 Sweeps
9 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/9]
      BatReturn  [SP=2]
   SollcScav 12
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 18:02, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
10 Traum     :  0.195     297.01 K
10 Tzelle    :  3.788     325.57 K     300 K
10 Tbad      :  0.001     272.18 K
10 TOHq      :  2.925     290.62 K    19.9971 mbar
10 TReakt    :  3.438     310.85 K     0.5227 mbar
10 pSat      :  7.465    1007.78 mbar
10 pZelle1000:  0.963     127.12 mbar   133 mbar
10 pZelle10  :  0.000       0.00 mbar
10 Flow1     :  2.940    1253.49 sccm
10 Flow2     :  4.121      45.67 sccm     Wasser:     2006.90     2000.00
10 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
10 Flow4     :  3.803       2.76 sccm         NO:       12.00       12.00
10 Flow5     : -0.003       0.00 sccm
10       sumflow   :     1302.85 sccm  1300.00 sccm
10       dosierFlow:     1302.85 sccm  9999.00 sccm
10       flowZelle :       10382  ccm        0  ccm
10 4096*Dwell= 5.001 s    100 Sweeps
10 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/10]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 18:18, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
11 Traum     :  0.195     297.01 K
11 Tzelle    :  3.788     325.58 K     300 K
11 Tbad      :  0.001     272.18 K
11 TOHq      :  2.924     290.61 K    19.9832 mbar
11 TReakt    :  3.438     310.85 K     0.5229 mbar
11 pSat      :  7.462    1007.41 mbar
11 pZelle1000:  0.964     127.19 mbar   133 mbar
11 pZelle10  :  0.000       0.00 mbar
11 Flow1     :  2.941    1253.68 sccm
11 Flow2     :  4.121      45.66 sccm     Wasser:     2005.84     2000.00
11 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
11 Flow4     :  4.784       3.46 sccm         NO:       15.01       15.00
11 Flow5     : -0.003       0.00 sccm
11       sumflow   :     1303.72 sccm  1300.00 sccm
11       dosierFlow:     1303.72 sccm  9999.00 sccm
11       flowZelle :       10383  ccm        0  ccm
11 4096*Dwell= 5.001 s    100 Sweeps
11 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/11]
      BatReturn  [SP=2]
   WartDauer 1200
   SolltEnd 3 [4096*0.732 ms]
   SollcReakt 3
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 18:50, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
12 Traum     :  0.195     297.00 K
12 Tzelle    :  3.789     325.59 K     300 K
12 Tbad      :  0.001     272.13 K
12 TOHq      :  2.924     290.60 K    19.9675 mbar
12 TReakt    :  3.438     310.85 K     0.5231 mbar
12 pSat      :  7.459    1006.95 mbar
12 pZelle1000:  0.963     127.18 mbar   133 mbar
12 pZelle10  :  0.000       0.00 mbar
12 Flow1     :  2.936    1251.51 sccm
12 Flow2     :  4.118      45.63 sccm     Wasser:     2008.15     2000.00
12 Flow3     :  2.309       2.69 sccm   m-Cresol:        3.04        3.00
12 Flow4     :  0.030       0.00 sccm         NO:        0.00        0.00
12 Flow5     :  0.000       0.00 sccm
12       sumflow   :     1300.75 sccm  1300.00 sccm
12       dosierFlow:     1300.75 sccm  9999.00 sccm
12       flowZelle :       10361  ccm        0  ccm
12 4096*Dwell= 2.998 s    100 Sweeps
12 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/12]
      BatReturn  [SP=2]
   WartDauer 300
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 19:03, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
13 Traum     :  0.196     297.13 K
13 Tzelle    :  3.789     325.61 K     300 K
13 Tbad      :  0.004     272.44 K
13 TOHq      :  2.925     290.62 K    19.9985 mbar
13 TReakt    :  3.439     310.89 K     0.5243 mbar
13 pSat      :  7.460    1007.06 mbar
13 pZelle1000:  0.970     128.03 mbar   133 mbar
13 pZelle10  :  0.000       0.00 mbar
13 Flow1     :  2.936    1251.68 sccm
13 Flow2     :  4.076      45.17 sccm     Wasser:     2004.11     2000.00
13 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.01        3.00
13 Flow4     :  0.448       0.25 sccm         NO:        1.09        1.00
13 Flow5     :  0.000       0.00 sccm
13       sumflow   :     1300.66 sccm  1300.00 sccm
13       dosierFlow:     1300.66 sccm  9999.00 sccm
13       flowZelle :       10291  ccm        0  ccm
13 4096*Dwell= 2.998 s    100 Sweeps
13 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/13]
      BatReturn  [SP=2]
   SollcScav 2
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 19:16, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
14 Traum     :  0.196     297.08 K
14 Tzelle    :  3.789     325.60 K     300 K
14 Tbad      :  0.003     272.43 K
14 TOHq      :  2.925     290.61 K    19.9861 mbar
14 TReakt    :  3.439     310.87 K     0.5239 mbar
14 pSat      :  7.459    1006.94 mbar
14 pZelle1000:  0.969     127.91 mbar   133 mbar
14 pZelle10  :  0.000       0.00 mbar
14 Flow1     :  2.937    1252.27 sccm
14 Flow2     :  4.076      45.17 sccm     Wasser:     2000.01     2000.00
14 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.01        3.00
14 Flow4     :  0.741       0.48 sccm         NO:        2.09        2.00
14 Flow5     : -0.001       0.00 sccm
14       sumflow   :     1301.48 sccm  1300.00 sccm
14       dosierFlow:     1301.48 sccm  9999.00 sccm
14       flowZelle :       10307  ccm        0  ccm
14 4096*Dwell= 2.998 s    100 Sweeps
14 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/14]
      BatReturn  [SP=2]
   SollcScav 3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 19:29, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
15 Traum     :  0.196     297.09 K
15 Tzelle    :  3.789     325.61 K     300 K
15 Tbad      :  0.003     272.43 K
15 TOHq      :  2.924     290.60 K    19.9723 mbar
15 TReakt    :  3.439     310.88 K     0.5240 mbar
15 pSat      :  7.459    1006.95 mbar
15 pZelle1000:  0.969     127.94 mbar   133 mbar
15 pZelle10  :  0.000       0.00 mbar
15 Flow1     :  2.938    1252.57 sccm
15 Flow2     :  4.078      45.19 sccm     Wasser:     1998.94     2000.00
15 Flow3     :  2.269       2.65 sccm   m-Cresol:        3.01        3.00
15 Flow4     :  1.019       0.69 sccm         NO:        3.02        3.00
15 Flow5     :  0.000       0.00 sccm
15       sumflow   :     1302.01 sccm  1300.00 sccm
15       dosierFlow:     1302.01 sccm  9999.00 sccm
15       flowZelle :       10309  ccm        0  ccm
15 4096*Dwell= 2.998 s    100 Sweeps
15 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/15]
      BatReturn  [SP=2]
   SollcScav 5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 19:42, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
16 Traum     :  0.195     297.02 K
16 Tzelle    :  3.789     325.60 K     300 K
16 Tbad      :  0.004     272.45 K
16 TOHq      :  2.924     290.59 K    19.9632 mbar
16 TReakt    :  3.439     310.87 K     0.5238 mbar
16 pSat      :  7.459    1006.97 mbar
16 pZelle1000:  0.969     127.95 mbar   133 mbar
16 pZelle10  :  0.000       0.00 mbar
16 Flow1     :  2.939    1252.80 sccm
16 Flow2     :  4.076      45.16 sccm     Wasser:     1996.10     2000.00
16 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.01        3.00
16 Flow4     :  1.635       1.16 sccm         NO:        5.08        5.00
16 Flow5     : -0.000       0.00 sccm
16       sumflow   :     1302.69 sccm  1300.00 sccm
16       dosierFlow:     1302.69 sccm  9999.00 sccm
16       flowZelle :       10313  ccm        0  ccm
16 4096*Dwell= 2.998 s    100 Sweeps
16 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/16]
      BatReturn  [SP=2]
   SollcScav 8
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 19:55, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
17 Traum     :  0.196     297.12 K
17 Tzelle    :  3.788     325.58 K     300 K
17 Tbad      :  0.003     272.43 K
17 TOHq      :  2.924     290.58 K    19.9451 mbar
17 TReakt    :  3.439     310.87 K     0.5238 mbar
17 pSat      :  7.458    1006.89 mbar
17 pZelle1000:  0.969     127.96 mbar   133 mbar
17 pZelle10  :  0.000       0.00 mbar
17 Flow1     :  2.939    1252.86 sccm
17 Flow2     :  4.077      45.18 sccm     Wasser:     1994.14     2000.00
17 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.01        3.00
17 Flow4     :  2.534       1.84 sccm         NO:        8.03        8.00
17 Flow5     : -0.000       0.00 sccm
17       sumflow   :     1303.44 sccm  1300.00 sccm
17       dosierFlow:     1303.44 sccm  9999.00 sccm
17       flowZelle :       10319  ccm        0  ccm
17 4096*Dwell= 2.998 s    100 Sweeps
17 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/17]
      BatReturn  [SP=2]
   SollcScav 10
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 20:08, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
18 Traum     :  0.196     297.13 K
18 Tzelle    :  3.788     325.58 K     300 K
18 Tbad      :  0.003     272.42 K
18 TOHq      :  2.924     290.59 K    19.9551 mbar
18 TReakt    :  3.439     310.87 K     0.5239 mbar
18 pSat      :  7.458    1006.82 mbar
18 pZelle1000:  0.969     127.92 mbar   133 mbar
18 pZelle10  :  0.000       0.00 mbar
18 Flow1     :  2.938    1252.44 sccm
18 Flow2     :  4.077      45.18 sccm     Wasser:     1994.68     2000.00
18 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.00        3.00
18 Flow4     :  3.148       2.29 sccm         NO:       10.00       10.00
18 Flow5     : -0.000       0.00 sccm
18       sumflow   :     1303.47 sccm  1300.00 sccm
18       dosierFlow:     1303.47 sccm  9999.00 sccm
18       flowZelle :       10322  ccm        0  ccm
18 4096*Dwell= 2.998 s    100 Sweeps
18 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/18]
      BatReturn  [SP=2]
   SollcScav 12
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 20:21, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
19 Traum     :  0.196     297.12 K
19 Tzelle    :  3.789     325.59 K     300 K
19 Tbad      :  0.003     272.43 K
19 TOHq      :  2.924     290.58 K    19.9503 mbar
19 TReakt    :  3.439     310.88 K     0.5241 mbar
19 pSat      :  7.457    1006.68 mbar
19 pZelle1000:  0.968     127.82 mbar   133 mbar
19 pZelle10  :  0.000       0.00 mbar
19 Flow1     :  2.937    1252.30 sccm
19 Flow2     :  4.118      45.63 sccm     Wasser:     2011.40     2000.00
19 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.00        3.00
19 Flow4     :  3.802       2.76 sccm         NO:       12.05       12.00
19 Flow5     :  0.000       0.00 sccm
19       sumflow   :     1304.25 sccm  1300.00 sccm
19       dosierFlow:     1304.25 sccm  9999.00 sccm
19       flowZelle :       10336  ccm        0  ccm
19 4096*Dwell= 2.998 s    100 Sweeps
19 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/19]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 20:34, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
20 Traum     :  0.196     297.11 K
20 Tzelle    :  3.788     325.58 K     300 K
20 Tbad      :  0.003     272.42 K
20 TOHq      :  2.924     290.58 K    19.9484 mbar
20 TReakt    :  3.439     310.88 K     0.5242 mbar
20 pSat      :  7.456    1006.53 mbar
20 pZelle1000:  0.969     127.97 mbar   133 mbar
20 pZelle10  :  0.000       0.00 mbar
20 Flow1     :  2.936    1251.93 sccm
20 Flow2     :  4.118      45.63 sccm     Wasser:     2013.33     2000.00
20 Flow3     :  2.268       2.64 sccm   m-Cresol:        3.01        3.00
20 Flow4     :  4.784       3.46 sccm         NO:       15.09       15.00
20 Flow5     :  0.000       0.00 sccm
20       sumflow   :     1304.58 sccm  1300.00 sccm
20       dosierFlow:     1304.58 sccm  9999.00 sccm
20       flowZelle :       10327  ccm        0  ccm
20 4096*Dwell= 2.998 s    100 Sweeps
20 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/20]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 20:47, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
21 Traum     :  0.196     297.13 K
21 Tzelle    :  3.789     325.59 K     300 K
21 Tbad      :  0.003     272.42 K
21 TOHq      :  2.924     290.58 K    19.9508 mbar
21 TReakt    :  3.439     310.88 K     0.5241 mbar
21 pSat      :  7.454    1006.32 mbar
21 pZelle1000:  0.969     127.92 mbar   133 mbar
21 pZelle10  :  0.000       0.00 mbar
21 Flow1     :  2.937    1252.31 sccm
21 Flow2     :  4.076      45.17 sccm     Wasser:     1993.13     2000.00
21 Flow3     :  2.268       2.65 sccm   m-Cresol:        3.01        3.00
21 Flow4     :  4.784       3.46 sccm         NO:       15.09       15.00
21 Flow5     : -0.000       0.00 sccm
21       sumflow   :     1304.50 sccm  1300.00 sccm
21       dosierFlow:     1304.50 sccm  9999.00 sccm
21       flowZelle :       10330  ccm        0  ccm
21 4096*Dwell= 2.998 s    100 Sweeps
21 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/21]
      BatReturn  [SP=2]
   SollcScav 12
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 21:00, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
22 Traum     :  0.196     297.08 K
22 Tzelle    :  3.789     325.59 K     300 K
22 Tbad      :  0.003     272.42 K
22 TOHq      :  2.924     290.58 K    19.9494 mbar
22 TReakt    :  3.439     310.88 K     0.5241 mbar
22 pSat      :  7.453    1006.15 mbar
22 pZelle1000:  0.969     127.91 mbar   133 mbar
22 pZelle10  :  0.000       0.00 mbar
22 Flow1     :  2.937    1252.10 sccm
22 Flow2     :  4.077      45.18 sccm     Wasser:     1995.11     2000.00
22 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.01        3.00
22 Flow4     :  3.802       2.76 sccm         NO:       12.06       12.00
22 Flow5     : -0.000       0.00 sccm
22       sumflow   :     1303.60 sccm  1300.00 sccm
22       dosierFlow:     1303.60 sccm  9999.00 sccm
22       flowZelle :       10324  ccm        0  ccm
22 4096*Dwell= 2.998 s    100 Sweeps
22 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/22]
      BatReturn  [SP=2]
   SollcScav 10
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 21:13, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
23 Traum     :  0.195     297.00 K
23 Tzelle    :  3.789     325.59 K     300 K
23 Tbad      :  0.003     272.41 K
23 TOHq      :  2.924     290.57 K    19.9408 mbar
23 TReakt    :  3.439     310.88 K     0.5242 mbar
23 pSat      :  7.452    1006.01 mbar
23 pZelle1000:  0.969     127.95 mbar   133 mbar
23 pZelle10  :  0.000       0.00 mbar
23 Flow1     :  2.936    1251.90 sccm
23 Flow2     :  4.077      45.18 sccm     Wasser:     1996.03     2000.00
23 Flow3     :  2.268       2.64 sccm   m-Cresol:        3.01        3.00
23 Flow4     :  3.149       2.29 sccm         NO:       10.01       10.00
23 Flow5     :  0.000       0.00 sccm
23       sumflow   :     1302.93 sccm  1300.00 sccm
23       dosierFlow:     1302.93 sccm  9999.00 sccm
23       flowZelle :       10316  ccm        0  ccm
23 4096*Dwell= 2.998 s    100 Sweeps
23 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/23]
      BatReturn  [SP=2]
   SollcScav 8
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 21:26, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
24 Traum     :  0.197     297.18 K
24 Tzelle    :  3.789     325.60 K     300 K
24 Tbad      :  0.004     272.44 K
24 TOHq      :  2.924     290.58 K    19.9451 mbar
24 TReakt    :  3.439     310.88 K     0.5240 mbar
24 pSat      :  7.451    1005.82 mbar
24 pZelle1000:  0.969     127.87 mbar   133 mbar
24 pZelle10  :  0.000       0.00 mbar
24 Flow1     :  2.938    1252.43 sccm
24 Flow2     :  4.075      45.16 sccm     Wasser:     1994.55     2000.00
24 Flow3     :  2.268       2.65 sccm   m-Cresol:        3.01        3.00
24 Flow4     :  2.534       1.84 sccm         NO:        8.03        8.00
24 Flow5     :  0.000       0.00 sccm
24       sumflow   :     1302.98 sccm  1300.00 sccm
24       dosierFlow:     1302.98 sccm  9999.00 sccm
24       flowZelle :       10322  ccm        0  ccm
24 4096*Dwell= 2.998 s    100 Sweeps
24 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/24]
      BatReturn  [SP=2]
   SollcScav 5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 21:39, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
25 Traum     :  0.196     297.05 K
25 Tzelle    :  3.789     325.60 K     300 K
25 Tbad      :  0.003     272.43 K
25 TOHq      :  2.924     290.60 K    19.9670 mbar
25 TReakt    :  3.439     310.88 K     0.5239 mbar
25 pSat      :  7.449    1005.67 mbar
25 pZelle1000:  0.969     127.92 mbar   133 mbar
25 pZelle10  :  0.000       0.00 mbar
25 Flow1     :  2.939    1252.79 sccm
25 Flow2     :  4.077      45.18 sccm     Wasser:     1999.45     2000.00
25 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.01        3.00
25 Flow4     :  1.636       1.16 sccm         NO:        5.08        5.00
25 Flow5     :  0.000       0.00 sccm
25       sumflow   :     1302.70 sccm  1300.00 sccm
25       dosierFlow:     1302.70 sccm  9999.00 sccm
25       flowZelle :       10316  ccm        0  ccm
25 4096*Dwell= 2.998 s    100 Sweeps
25 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/25]
      BatReturn  [SP=2]
   SollcScav 3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 21:52, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
26 Traum     :  0.196     297.09 K
26 Tzelle    :  3.789     325.60 K     300 K
26 Tbad      :  0.004     272.44 K
26 TOHq      :  2.924     290.59 K    19.9618 mbar
26 TReakt    :  3.439     310.88 K     0.5242 mbar
26 pSat      :  7.449    1005.55 mbar
26 pZelle1000:  0.970     128.01 mbar   133 mbar
26 pZelle10  :  0.000       0.00 mbar
26 Flow1     :  2.938    1252.45 sccm
26 Flow2     :  4.073      45.14 sccm     Wasser:     1999.83     2000.00
26 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.02        3.00
26 Flow4     :  1.025       0.70 sccm         NO:        3.05        3.00
26 Flow5     : -0.000       0.00 sccm
26       sumflow   :     1301.84 sccm  1300.00 sccm
26       dosierFlow:     1301.84 sccm  9999.00 sccm
26       flowZelle :       10302  ccm        0  ccm
26 4096*Dwell= 2.998 s    100 Sweeps
26 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/26]
      BatReturn  [SP=2]
   SollcScav 2
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 22:05, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
27 Traum     :  0.197     297.17 K
27 Tzelle    :  3.789     325.59 K     300 K
27 Tbad      :  0.004     272.46 K
27 TOHq      :  2.924     290.60 K    19.9785 mbar
27 TReakt    :  3.439     310.88 K     0.5241 mbar
27 pSat      :  7.447    1005.32 mbar
27 pZelle1000:  0.969     127.94 mbar   133 mbar
27 pZelle10  :  0.000       0.00 mbar
27 Flow1     :  2.936    1251.65 sccm
27 Flow2     :  4.078      45.19 sccm     Wasser:     2004.59     2000.00
27 Flow3     :  2.268       2.64 sccm   m-Cresol:        3.02        3.00
27 Flow4     :  0.737       0.47 sccm         NO:        2.07        2.00
27 Flow5     : -0.001       0.00 sccm
27       sumflow   :     1300.87 sccm  1300.00 sccm
27       dosierFlow:     1300.87 sccm  9999.00 sccm
27       flowZelle :       10300  ccm        0  ccm
27 4096*Dwell= 2.998 s    100 Sweeps
27 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/27]
      BatReturn  [SP=2]
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 22:18, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
28 Traum     :  0.196     297.11 K
28 Tzelle    :  3.789     325.59 K     300 K
28 Tbad      :  0.004     272.48 K
28 TOHq      :  2.924     290.60 K    19.9751 mbar
28 TReakt    :  3.439     310.88 K     0.5239 mbar
28 pSat      :  7.445    1005.13 mbar
28 pZelle1000:  0.968     127.84 mbar   133 mbar
28 pZelle10  :  0.000       0.00 mbar
28 Flow1     :  2.937    1252.07 sccm
28 Flow2     :  4.076      45.16 sccm     Wasser:     2001.81     2000.00
28 Flow3     :  2.267       2.64 sccm   m-Cresol:        3.01        3.00
28 Flow4     :  0.449       0.25 sccm         NO:        1.09        1.00
28 Flow5     : -0.000       0.00 sccm
28       sumflow   :     1301.04 sccm  1300.00 sccm
28       dosierFlow:     1301.04 sccm  9999.00 sccm
28       flowZelle :       10309  ccm        0  ccm
28 4096*Dwell= 2.998 s    100 Sweeps
28 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/28]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 22:33, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
29 Traum     :  0.195     297.03 K
29 Tzelle    :  3.787     325.54 K     300 K
29 Tbad      :  0.001     272.12 K
29 TOHq      :  2.923     290.56 K    19.9198 mbar
29 TReakt    :  3.438     310.85 K     0.5229 mbar
29 pSat      :  7.442    1004.67 mbar
29 pZelle1000:  0.963     127.15 mbar   133 mbar
29 pZelle10  :  0.000       0.00 mbar
29 Flow1     :  2.935    1251.42 sccm
29 Flow2     :  4.118      45.63 sccm     Wasser:     2007.87     2000.00
29 Flow3     :  2.308       2.69 sccm   m-Cresol:        3.04        3.00
29 Flow4     :  0.026       0.00 sccm         NO:        0.00        0.00
29 Flow5     :  0.001       0.00 sccm
29       sumflow   :     1300.66 sccm  1300.00 sccm
29       dosierFlow:     1300.66 sccm  9999.00 sccm
29       flowZelle :       10362  ccm        0  ccm
29 4096*Dwell= 2.998 s    100 Sweeps
29 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/29]
      BatReturn  [SP=2]
   WartDauer 1200
   SolltEnd 3 [4096*0.732 ms]
   SollcReakt 6
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 23:01, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
30 Traum     :  0.195     296.99 K
30 Tzelle    :  3.788     325.55 K     300 K
30 Tbad      :  0.001     272.10 K
30 TOHq      :  2.923     290.55 K    19.9060 mbar
30 TReakt    :  3.438     310.85 K     0.5229 mbar
30 pSat      :  7.440    1004.45 mbar
30 pZelle1000:  0.963     127.16 mbar   133 mbar
30 pZelle10  :  0.000       0.00 mbar
30 Flow1     :  2.936    1251.93 sccm
30 Flow2     :  4.120      45.65 sccm     Wasser:     2003.17     2000.00
30 Flow3     :  5.003       5.26 sccm   m-Cresol:        5.95        6.00
30 Flow4     :  0.032       0.00 sccm         NO:        0.00        0.00
30 Flow5     : -0.002       0.00 sccm
30       sumflow   :     1303.78 sccm  1300.00 sccm
30       dosierFlow:     1303.78 sccm  9999.00 sccm
30       flowZelle :       10386  ccm        0  ccm
30 4096*Dwell= 2.998 s    100 Sweeps
30 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/30]
      BatReturn  [SP=2]
   WartDauer 300
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 23:14, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
31 Traum     :  0.195     296.99 K
31 Tzelle    :  3.788     325.57 K     300 K
31 Tbad      :  0.004     272.44 K
31 TOHq      :  2.924     290.60 K    19.9737 mbar
31 TReakt    :  3.439     310.90 K     0.5247 mbar
31 pSat      :  7.441    1004.48 mbar
31 pZelle1000:  0.969     127.91 mbar   133 mbar
31 pZelle10  :  0.000       0.00 mbar
31 Flow1     :  2.937    1252.09 sccm
31 Flow2     :  4.079      45.20 sccm     Wasser:     2001.55     2000.00
31 Flow3     :  5.003       5.27 sccm   m-Cresol:        6.01        6.00
31 Flow4     :  0.451       0.25 sccm         NO:        1.09        1.00
31 Flow5     : -0.001       0.00 sccm
31       sumflow   :     1303.72 sccm  1300.00 sccm
31       dosierFlow:     1303.72 sccm  9999.00 sccm
31       flowZelle :       10325  ccm        0  ccm
31 4096*Dwell= 2.998 s    100 Sweeps
31 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/31]
      BatReturn  [SP=2]
   SollcScav 2
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 23:27, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
32 Traum     :  0.195     297.02 K
32 Tzelle    :  3.788     325.57 K     300 K
32 Tbad      :  0.003     272.42 K
32 TOHq      :  2.924     290.59 K    19.9627 mbar
32 TReakt    :  3.439     310.87 K     0.5239 mbar
32 pSat      :  7.440    1004.43 mbar
32 pZelle1000:  0.970     128.03 mbar   133 mbar
32 pZelle10  :  0.000       0.00 mbar
32 Flow1     :  2.938    1252.70 sccm
32 Flow2     :  4.077      45.18 sccm     Wasser:     2000.52     2000.00
32 Flow3     :  5.003       5.26 sccm   m-Cresol:        6.00        6.00
32 Flow4     :  0.741       0.48 sccm         NO:        2.08        2.00
32 Flow5     : -0.001       0.00 sccm
32       sumflow   :     1304.54 sccm  1300.00 sccm
32       dosierFlow:     1304.54 sccm  9999.00 sccm
32       flowZelle :       10322  ccm        0  ccm
32 4096*Dwell= 2.998 s    100 Sweeps
32 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/32]
      BatReturn  [SP=2]
   SollcScav 3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 23:40, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
33 Traum     :  0.196     297.06 K
33 Tzelle    :  3.789     325.58 K     300 K
33 Tbad      :  0.003     272.43 K
33 TOHq      :  2.924     290.59 K    19.9651 mbar
33 TReakt    :  3.439     310.88 K     0.5240 mbar
33 pSat      :  7.439    1004.30 mbar
33 pZelle1000:  0.969     127.90 mbar   133 mbar
33 pZelle10  :  0.000       0.00 mbar
33 Flow1     :  2.937    1252.12 sccm
33 Flow2     :  4.078      45.19 sccm     Wasser:     2000.00     2000.00
33 Flow3     :  5.002       5.26 sccm   m-Cresol:        6.00        6.00
33 Flow4     :  1.026       0.70 sccm         NO:        3.04        3.00
33 Flow5     : -0.000       0.00 sccm
33       sumflow   :     1304.20 sccm  1300.00 sccm
33       dosierFlow:     1304.20 sccm  9999.00 sccm
33       flowZelle :       10329  ccm        0  ccm
33 4096*Dwell= 2.998 s    100 Sweeps
33 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/33]
      BatReturn  [SP=2]
   SollcScav 5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 2.9. 23:53, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
34 Traum     :  0.196     297.16 K
34 Tzelle    :  3.789     325.58 K     300 K
34 Tbad      :  0.003     272.43 K
34 TOHq      :  2.924     290.59 K    19.9541 mbar
34 TReakt    :  3.439     310.89 K     0.5244 mbar
34 pSat      :  7.439    1004.25 mbar
34 pZelle1000:  0.969     127.93 mbar   133 mbar
34 pZelle10  :  0.000       0.00 mbar
34 Flow1     :  2.936    1251.78 sccm
34 Flow2     :  4.079      45.20 sccm     Wasser:     1999.60     2000.00
34 Flow3     :  5.003       5.26 sccm   m-Cresol:        6.00        6.00
34 Flow4     :  1.634       1.16 sccm         NO:        5.08        5.00
34 Flow5     :  0.000       0.00 sccm
34       sumflow   :     1304.33 sccm  1300.00 sccm
34       dosierFlow:     1304.33 sccm  9999.00 sccm
34       flowZelle :       10328  ccm        0  ccm
34 4096*Dwell= 2.998 s    100 Sweeps
34 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/34]
      BatReturn  [SP=2]
   SollcScav 8
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 0:06, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
35 Traum     :  0.196     297.07 K
35 Tzelle    :  3.789     325.61 K     300 K
35 Tbad      :  0.003     272.42 K
35 TOHq      :  2.924     290.58 K    19.9527 mbar
35 TReakt    :  3.439     310.89 K     0.5245 mbar
35 pSat      :  7.438    1004.09 mbar
35 pZelle1000:  0.970     127.98 mbar   133 mbar
35 pZelle10  :  0.000       0.00 mbar
35 Flow1     :  2.937    1252.32 sccm
35 Flow2     :  4.079      45.20 sccm     Wasser:     1998.46     2000.00
35 Flow3     :  4.995       5.26 sccm   m-Cresol:        5.99        6.00
35 Flow4     :  2.534       1.84 sccm         NO:        8.02        8.00
35 Flow5     : -0.001       0.00 sccm
35       sumflow   :     1305.54 sccm  1300.00 sccm
35       dosierFlow:     1305.54 sccm  9999.00 sccm
35       flowZelle :       10334  ccm        0  ccm
35 4096*Dwell= 2.998 s    100 Sweeps
35 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/35]
      BatReturn  [SP=2]
   SollcScav 10
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 0:18, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
36 Traum     :  0.196     297.10 K
36 Tzelle    :  3.789     325.59 K     300 K
36 Tbad      :  0.003     272.42 K
36 TOHq      :  2.924     290.58 K    19.9508 mbar
36 TReakt    :  3.439     310.88 K     0.5241 mbar
36 pSat      :  7.437    1003.99 mbar
36 pZelle1000:  0.969     127.85 mbar   133 mbar
36 pZelle10  :  0.000       0.00 mbar
36 Flow1     :  2.937    1252.23 sccm
36 Flow2     :  4.079      45.20 sccm     Wasser:     1996.08     2000.00
36 Flow3     :  5.007       5.27 sccm   m-Cresol:        5.99        6.00
36 Flow4     :  3.190       2.32 sccm         NO:       10.11       10.00
36 Flow5     : -0.000       0.00 sccm
36       sumflow   :     1305.94 sccm  1300.00 sccm
36       dosierFlow:     1305.94 sccm  9999.00 sccm
36       flowZelle :       10348  ccm        0  ccm
36 4096*Dwell= 2.998 s    100 Sweeps
36 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/36]
      BatReturn  [SP=2]
   SollcScav 12
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 0:31, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
37 Traum     :  0.195     297.03 K
37 Tzelle    :  3.788     325.58 K     300 K
37 Tbad      :  0.003     272.41 K
37 TOHq      :  2.924     290.58 K    19.9498 mbar
37 TReakt    :  3.439     310.88 K     0.5242 mbar
37 pSat      :  7.435    1003.79 mbar
37 pZelle1000:  0.970     128.06 mbar   133 mbar
37 pZelle10  :  0.000       0.00 mbar
37 Flow1     :  2.938    1252.45 sccm
37 Flow2     :  4.080      45.21 sccm     Wasser:     1999.07     2000.00
37 Flow3     :  5.003       5.26 sccm   m-Cresol:        6.00        6.00
37 Flow4     :  3.802       2.76 sccm         NO:       12.05       12.00
37 Flow5     : -0.000       0.00 sccm
37       sumflow   :     1306.60 sccm  1300.00 sccm
37       dosierFlow:     1306.60 sccm  9999.00 sccm
37       flowZelle :       10336  ccm        0  ccm
37 4096*Dwell= 2.998 s    100 Sweeps
37 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/37]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 0:44, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
38 Traum     :  0.195     297.01 K
38 Tzelle    :  3.788     325.58 K     300 K
38 Tbad      :  0.003     272.41 K
38 TOHq      :  2.924     290.59 K    19.9536 mbar
38 TReakt    :  3.439     310.88 K     0.5239 mbar
38 pSat      :  7.435    1003.68 mbar
38 pZelle1000:  0.968     127.82 mbar   133 mbar
38 pZelle10  :  0.000       0.00 mbar
38 Flow1     :  2.937    1252.22 sccm
38 Flow2     :  4.080      45.21 sccm     Wasser:     1995.34     2000.00
38 Flow3     :  5.001       5.26 sccm   m-Cresol:        5.98        6.00
38 Flow4     :  4.784       3.46 sccm         NO:       15.05       15.00
38 Flow5     : -0.001       0.00 sccm
38       sumflow   :     1307.08 sccm  1300.00 sccm
38       dosierFlow:     1307.08 sccm  9999.00 sccm
38       flowZelle :       10359  ccm        0  ccm
38 4096*Dwell= 2.998 s    100 Sweeps
38 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/38]
      BatReturn  [SP=2]
   SollcScav 15
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 0:57, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
39 Traum     :  0.195     297.01 K
39 Tzelle    :  3.788     325.57 K     300 K
39 Tbad      :  0.003     272.41 K
39 TOHq      :  2.924     290.59 K    19.9570 mbar
39 TReakt    :  3.439     310.88 K     0.5242 mbar
39 pSat      :  7.434    1003.62 mbar
39 pZelle1000:  0.969     127.88 mbar   133 mbar
39 pZelle10  :  0.000       0.00 mbar
39 Flow1     :  2.937    1252.29 sccm
39 Flow2     :  4.080      45.21 sccm     Wasser:     1996.57     2000.00
39 Flow3     :  5.003       5.26 sccm   m-Cresol:        5.99        6.00
39 Flow4     :  4.785       3.46 sccm         NO:       15.06       15.00
39 Flow5     : -0.000       0.00 sccm
39       sumflow   :     1307.14 sccm  1300.00 sccm
39       dosierFlow:     1307.14 sccm  9999.00 sccm
39       flowZelle :       10355  ccm        0  ccm
39 4096*Dwell= 2.998 s    100 Sweeps
39 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/39]
      BatReturn  [SP=2]
   SollcScav 12
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 1:10, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
40 Traum     :  0.197     297.20 K
40 Tzelle    :  3.788     325.58 K     300 K
40 Tbad      :  0.003     272.40 K
40 TOHq      :  2.924     290.59 K    19.9546 mbar
40 TReakt    :  3.439     310.88 K     0.5239 mbar
40 pSat      :  7.433    1003.39 mbar
40 pZelle1000:  0.969     127.85 mbar   133 mbar
40 pZelle10  :  0.000       0.00 mbar
40 Flow1     :  2.938    1252.39 sccm
40 Flow2     :  4.080      45.21 sccm     Wasser:     1997.25     2000.00
40 Flow3     :  5.003       5.26 sccm   m-Cresol:        5.99        6.00
40 Flow4     :  3.802       2.76 sccm         NO:       12.03       12.00
40 Flow5     : -0.000       0.00 sccm
40       sumflow   :     1306.55 sccm  1300.00 sccm
40       dosierFlow:     1306.55 sccm  9999.00 sccm
40       flowZelle :       10353  ccm        0  ccm
40 4096*Dwell= 2.998 s    100 Sweeps
40 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/40]
      BatReturn  [SP=2]
   SollcScav 10
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 1:23, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
41 Traum     :  0.196     297.15 K
41 Tzelle    :  3.789     325.60 K     300 K
41 Tbad      :  0.003     272.41 K
41 TOHq      :  2.924     290.59 K    19.9551 mbar
41 TReakt    :  3.439     310.89 K     0.5244 mbar
41 pSat      :  7.432    1003.27 mbar
41 pZelle1000:  0.969     127.93 mbar   133 mbar
41 pZelle10  :  0.000       0.00 mbar
41 Flow1     :  2.937    1252.23 sccm
41 Flow2     :  4.080      45.21 sccm     Wasser:     1999.40     2000.00
41 Flow3     :  5.002       5.26 sccm   m-Cresol:        6.00        6.00
41 Flow4     :  3.190       2.32 sccm         NO:       10.11       10.00
41 Flow5     : -0.001       0.00 sccm
41       sumflow   :     1305.94 sccm  1300.00 sccm
41       dosierFlow:     1305.94 sccm  9999.00 sccm
41       flowZelle :       10341  ccm        0  ccm
41 4096*Dwell= 2.998 s    100 Sweeps
41 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/41]
      BatReturn  [SP=2]
   SollcScav 8
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 1:36, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
42 Traum     :  0.196     297.14 K
42 Tzelle    :  3.789     325.60 K     300 K
42 Tbad      :  0.003     272.43 K
42 TOHq      :  2.924     290.59 K    19.9594 mbar
42 TReakt    :  3.439     310.88 K     0.5241 mbar
42 pSat      :  7.431    1003.23 mbar
42 pZelle1000:  0.968     127.84 mbar   133 mbar
42 pZelle10  :  0.000       0.00 mbar
42 Flow1     :  2.937    1252.11 sccm
42 Flow2     :  4.080      45.21 sccm     Wasser:     1999.50     2000.00
42 Flow3     :  5.004       5.27 sccm   m-Cresol:        6.00        6.00
42 Flow4     :  2.534       1.84 sccm         NO:        8.01        8.00
42 Flow5     :  0.000       0.00 sccm
42       sumflow   :     1305.34 sccm  1300.00 sccm
42       dosierFlow:     1305.34 sccm  9999.00 sccm
42       flowZelle :       10343  ccm        0  ccm
42 4096*Dwell= 2.998 s    100 Sweeps
42 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/42]
      BatReturn  [SP=2]
   SollcScav 5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 1:49, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
43 Traum     :  0.197     297.19 K
43 Tzelle    :  3.788     325.58 K     300 K
43 Tbad      :  0.003     272.41 K
43 TOHq      :  2.924     290.60 K    19.9770 mbar
43 TReakt    :  3.439     310.88 K     0.5242 mbar
43 pSat      :  7.432    1003.31 mbar
43 pZelle1000:  0.969     127.87 mbar   133 mbar
43 pZelle10  :  0.000       0.00 mbar
43 Flow1     :  2.937    1251.94 sccm
43 Flow2     :  4.079      45.20 sccm     Wasser:     2002.72     2000.00
43 Flow3     :  5.004       5.27 sccm   m-Cresol:        6.00        6.00
43 Flow4     :  1.635       1.16 sccm         NO:        5.07        5.00
43 Flow5     :  0.001       0.00 sccm
43       sumflow   :     1304.50 sccm  1300.00 sccm
43       dosierFlow:     1304.50 sccm  9999.00 sccm
43       flowZelle :       10335  ccm        0  ccm
43 4096*Dwell= 2.998 s    100 Sweeps
43 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/43]
      BatReturn  [SP=2]
   SollcScav 3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 2:02, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
44 Traum     :  0.196     297.11 K
44 Tzelle    :  3.788     325.57 K     300 K
44 Tbad      :  0.003     272.42 K
44 TOHq      :  2.924     290.61 K    19.9827 mbar
44 TReakt    :  3.439     310.89 K     0.5243 mbar
44 pSat      :  7.431    1003.20 mbar
44 pZelle1000:  0.969     127.96 mbar   133 mbar
44 pZelle10  :  0.000       0.00 mbar
44 Flow1     :  2.937    1252.07 sccm
44 Flow2     :  4.079      45.20 sccm     Wasser:     2005.52     2000.00
44 Flow3     :  5.001       5.26 sccm   m-Cresol:        6.01        6.00
44 Flow4     :  1.049       0.72 sccm         NO:        3.12        3.00
44 Flow5     :  0.000       0.00 sccm
44       sumflow   :     1304.18 sccm  1300.00 sccm
44       dosierFlow:     1304.18 sccm  9999.00 sccm
44       flowZelle :       10325  ccm        0  ccm
44 4096*Dwell= 2.998 s    100 Sweeps
44 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/44]
      BatReturn  [SP=2]
   SollcScav 2
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 2:15, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
45 Traum     :  0.196     297.04 K
45 Tzelle    :  3.788     325.58 K     300 K
45 Tbad      :  0.004     272.44 K
45 TOHq      :  2.924     290.60 K    19.9699 mbar
45 TReakt    :  3.439     310.89 K     0.5245 mbar
45 pSat      :  7.430    1003.07 mbar
45 pZelle1000:  0.969     127.96 mbar   133 mbar
45 pZelle10  :  0.000       0.00 mbar
45 Flow1     :  2.937    1252.19 sccm
45 Flow2     :  4.080      45.21 sccm     Wasser:     2004.90     2000.00
45 Flow3     :  4.991       5.25 sccm   m-Cresol:        6.00        6.00
45 Flow4     :  0.703       0.45 sccm         NO:        1.95        2.00
45 Flow5     : -0.000       0.00 sccm
45       sumflow   :     1304.03 sccm  1300.00 sccm
45       dosierFlow:     1304.03 sccm  9999.00 sccm
45       flowZelle :       10323  ccm        0  ccm
45 4096*Dwell= 2.998 s    100 Sweeps
45 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/45]
      BatReturn  [SP=2]
   SollcScav 1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 2:28, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
46 Traum     :  0.195     297.02 K
46 Tzelle    :  3.788     325.56 K     300 K
46 Tbad      :  0.004     272.44 K
46 TOHq      :  2.924     290.60 K    19.9765 mbar
46 TReakt    :  3.439     310.89 K     0.5243 mbar
46 pSat      :  7.429    1002.85 mbar
46 pZelle1000:  0.969     127.89 mbar   133 mbar
46 pZelle10  :  0.000       0.00 mbar
46 Flow1     :  2.937    1252.17 sccm
46 Flow2     :  4.080      45.21 sccm     Wasser:     2005.17     2000.00
46 Flow3     :  5.003       5.26 sccm   m-Cresol:        6.01        6.00
46 Flow4     :  0.451       0.25 sccm         NO:        1.09        1.00
46 Flow5     : -0.001       0.00 sccm
46       sumflow   :     1303.81 sccm  1300.00 sccm
46       dosierFlow:     1303.81 sccm  9999.00 sccm
46       flowZelle :       10328  ccm        0  ccm
46 4096*Dwell= 2.998 s    100 Sweeps
46 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/46]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 2:43, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
47 Traum     :  0.195     297.01 K
47 Tzelle    :  3.787     325.53 K     300 K
47 Tbad      :  0.001     272.10 K
47 TOHq      :  2.923     290.54 K    19.8942 mbar
47 TReakt    :  3.438     310.85 K     0.5227 mbar
47 pSat      :  7.425    1002.36 mbar
47 pZelle1000:  0.963     127.08 mbar   133 mbar
47 pZelle10  :  0.000       0.00 mbar
47 Flow1     :  2.937    1252.09 sccm
47 Flow2     :  4.121      45.66 sccm     Wasser:     2005.06     2000.00
47 Flow3     :  5.003       5.26 sccm   m-Cresol:        5.96        6.00
47 Flow4     :  0.027       0.00 sccm         NO:        0.00        0.00
47 Flow5     : -0.000       0.00 sccm
47       sumflow   :     1303.94 sccm  1300.00 sccm
47       dosierFlow:     1303.94 sccm  9999.00 sccm
47       flowZelle :       10394  ccm        0  ccm
47 4096*Dwell= 2.998 s    100 Sweeps
47 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/47]
      BatReturn  [SP=2]
   WartDauer 1200
   SolltEnd 5 [4096*1.221 ms]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 3:14, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
48 Traum     :  0.195     296.95 K
48 Tzelle    :  3.786     325.49 K     300 K
48 Tbad      : -0.001     271.90 K
48 TOHq      :  2.922     290.52 K    19.8775 mbar
48 TReakt    :  3.437     310.82 K     0.5215 mbar
48 pSat      :  7.421    1001.82 mbar
48 pZelle1000:  0.957     126.35 mbar   133 mbar
48 pZelle10  :  0.000       0.00 mbar
48 Flow1     :  2.936    1251.68 sccm
48 Flow2     :  4.120      45.65 sccm     Wasser:     2001.46     2000.00
48 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
48 Flow4     :  0.034       0.00 sccm         NO:        0.00        0.00
48 Flow5     : -0.002       0.00 sccm
48       sumflow   :     1298.25 sccm  1300.00 sccm
48       dosierFlow:     1298.25 sccm  9999.00 sccm
48       flowZelle :       10409  ccm        0  ccm
48 4096*Dwell= 5.001 s    100 Sweeps
48 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/48]
      BatReturn  [SP=2]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 3.9. 3:45, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
49 Traum     :  0.195     296.94 K
49 Tzelle    :  3.785     325.46 K     300 K
49 Tbad      : -0.001     271.88 K
49 TOHq      :  2.922     290.52 K    19.8718 mbar
49 TReakt    :  3.437     310.81 K     0.5213 mbar
49 pSat      :  7.420    1001.66 mbar
49 pZelle1000:  0.958     126.41 mbar   133 mbar
49 pZelle10  :  0.000       0.00 mbar
49 Flow1     :  2.937    1252.33 sccm
49 Flow2     :  4.123      45.68 sccm     Wasser:     2002.77     2000.00
49 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
49 Flow4     :  0.036       0.00 sccm         NO:        0.00        0.00
49 Flow5     : -0.003       0.00 sccm
49       sumflow   :     1298.93 sccm  1300.00 sccm
49       dosierFlow:     1298.93 sccm  9999.00 sccm
49       flowZelle :       10409  ccm        0  ccm
49 4096*Dwell= 5.001 s    100 Sweeps
49 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [12/49]
      BatReturn  [SP=2]
   BatReturn  [SP=1]
DosBefehl :
