SaveOpen cresol7.dec 7 [bereit fr 7/1]
BatGosub std [SP=2]
   Inert Argon
   OHq Wasser
   Reakt m-Cresol
   Scav NO2 2.595e-4
   Reakt3 NC 0
   SollSumflow 1300
   SollFlowZelle 0 9999
   WartDauer 300
   SollcOHq 2000
   SollcReakt 0
   SollcScav 0
   SollcReakt3 190
   SollSweeps 50
   SollblFrequ 0.25
   SollBlEnergie 4
   SolltEnd 4 [4096*0.977 ms]
   SollTzelle 300
   SollpZelle 133
   SollTrohr 299
   BatReturn  [SP=1]
SollcReakt 6
SollcScav .05
EinstKonz 
ZeigeZeit  [23.8. 15:10]
Ende 
SaveOpen cresol7.dec 7 [bereit fr 7/1]
BatGosub std [SP=2]
   Inert Argon
   OHq Wasser
   Reakt m-Cresol
   Scav NO2 2.595e-4
   Reakt3 NC 0
   SollSumflow 1300
   SollFlowZelle 0 9999
   WartDauer 300
   SollcOHq 2000
   SollcReakt 0
   SollcScav 0
   SollcReakt3 190
   SollSweeps 50
   SollblFrequ 0.25
   SollBlEnergie 4
   SolltEnd 4 [4096*0.977 ms]
   SollTzelle 300
   SollpZelle 133
   SollTrohr 299
   BatReturn  [SP=1]
BatMsg  [tylan3 ausgetauscht: jetzt wieder 5sccm]
ZeigeZeit  [24.8. 15:11]
ZeigeAntoine [
  8.610  1948  -24.1    Wasser(0-19 Grad),
  8.150  2124  -59.0    m-cresol(C7H8O, St ]
ZeigeCalib [
        name   ad   da   io  offset   gain linear.    Bemerkung
       traum   16   -1   -1  273.50 120.400 0.0000  RTD(0!!!..600ø)
  tzellpt100   15   -1   -1  174.80  39.800 0.0000  RTD(-100..100ø)
        tbad   13   -1   -1  272.00 125.000 0.0000  K 20 -Thermostat
        tohq   17   -1   -1  174.80  39.600 0.0000  RTD(-100..100ø)
      treakt   18   -1   -1  174.70  39.600 0.0000  RTD(-100..100ø)
  pzelle1000    5   -1   -1    0.00 132.000 0.0000  MKS(0..1000mbar)
    pzelle10   -7   -1   -1   -0.30   1.381 0.0000  MKS(0..10torr)
  psetze1000   -1    4 1015    0.00 -266.600 0.0000  Setze Pzelle1000
    psetze10   -1    4  -15   -0.30  -2.000 0.0000  Setze Pzelle10
        psat    6   -1   -1    0.00 135.000 0.0000  MKS(0..1000mbar)
       flow1    0    0   -1   94.03 364.000 10.3230  Bo822527_2  2000Ar
       flow2    1    6    1    1.85  10.240 0.0950   Bo822530_5  50Ar
       flow3    2    8    0    0.12   1.186 -0.0314  A_192_84_1568 5Ar
       flow4    3   10    2    0.19  39.360 0.1321  A_192_89_0066 200Ar
       flow5    4    2   -1    0.08  46.910 -0.6557  A_192_82_4885 200Ar
     blitzn2   -1   -1    4    0.00   0.000 0.0000  Magnetventil
   bltrigger   -1   -1   12    0.00   0.000 0.0000  PC30 an Zndspule ]
ProtIstWerte [
             Spannung    Istwert
Traum     :  0.197     297.25 K
Tzelle    :  3.779     325.20 K     300 K
Tbad      : -0.001     271.83 K
TOHq      :  3.046     295.41 K    26.9120 mbar
TReakt    :  3.439     310.90 K     0.5249 mbar
pSat      :  7.466    1007.91 mbar
pZelle1000:  0.949     125.27 mbar   133 mbar
pZelle10  :  0.000       0.00 mbar
Flow1     :  2.935    1251.43 sccm
Flow2     :  3.021      33.65 sccm     Wasser:     1971.21     2000.00
Flow3     :  3.540       3.92 sccm   m-Cresol:        4.36        0.00
Flow4     :  4.338     173.43 sccm        NO2:        0.05        0.00
Flow5     :  4.311     190.12 sccm
      sumflow   :     1306.62 sccm  1300.00 sccm
      dosierFlow:     1116.50 sccm  9999.00 sccm
      flowZelle :       10566  ccm        0  ccm
4096*Dwell= 4.000 s    50 Sweeps
BlitzFolge= 4.000 s    BlitzEnergie  4.00 J
<any key> ]
  [Fehler: ADC-Overflow (>9.5V)]
SollcReakt 6
SollcScav .05
EinstKonz 
ProtIstWerte [
             Spannung    Istwert
Traum     :  0.200     297.52 K
Tzelle    :  3.781     325.28 K     300 K
Tbad      :  0.003     272.42 K
TOHq      :  3.049     295.53 K    27.1080 mbar
TReakt    :  3.442     311.01 K     0.5294 mbar
pSat      :  7.472    1008.70 mbar
pZelle1000:  0.967     127.67 mbar   133 mbar
pZelle10  :  0.000       0.00 mbar
Flow1     :  2.938    1252.47 sccm
Flow2     :  2.982      33.23 sccm     Wasser:     1993.62     2000.00
Flow3     :  5.008       5.27 sccm   m-Cresol:        6.01        6.00
Flow4     :  4.333     173.22 sccm        NO2:        0.05        0.05
Flow5     :  4.312     190.15 sccm
      sumflow   :     1308.82 sccm  1300.00 sccm
      dosierFlow:     1118.67 sccm  9999.00 sccm
      flowZelle :       10385  ccm        0  ccm
4096*Dwell= 4.000 s    50 Sweeps
BlitzFolge= 4.000 s    BlitzEnergie  4.00 J
<any key> ]
  [Fehler: ADC-Overflow (>9.5V)]
SollcScav 0
SollcReakt 0
EinstKonz 
SollBlEnergie 2.4
DosBefehl :t w.
[
SollcReakt 0
WartDauer  5
SollSweeps 50   SolltEnd 2
     SollcScav 0.5  Batgosub m
     SollcScav 0    BatGosub m
                    BatGosub m
                    BatGosub m
                    BatGosub m
                    BatGosub m
                    BatGosub m
                    BatGosub m
WartDauer 1200
                SolltEnd 5
SollcReakt    0    BatGosub m
SollcReakt    0    Batgosub m
WartDauer 600
Sollsweeps 100  SolltEnd 2
     SollcScav 0.05 BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.05 BatGosub m
     SollcScav 0    BatGosub m
     SollcScav 0    BatGosub m
WartDauer 1200
SollSweeps 200  SolltEnd 3
SollcReakt 1.0    BatGosub m
SollcReakt 2.0    BatGosub m
                SolltEnd 2
SollcReakt 3.0    BatGosub m
WartDauer 600
     SollcScav 0.05 BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.05 BatGosub m
     SollcScav 0    BatGosub m
     SollcScav 0    BatGosub m
WartDauer 1200
SollcReakt 4.0    BatGosub m
SollSweeps 300  SolltEnd 1
SollcReakt 5.0    BatGosub m
SollcReakt 6.0    BatGosub m
SollcReakt 6.0    BatGosub m
SollcReakt 5.0    BatGosub m
SollSweeps 200  SolltEnd 2
SollcReakt 4.0    BatGosub m
SollcReakt 3.0    BatGosub m
WartDauer 600
     SollcScav 0.05 BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.5  BatGosub m
     SollcScav 0.3  BatGosub m
     SollcScav 0.1  BatGosub m
     SollcScav 0.05 BatGosub m
     SollcScav 0    BatGosub m
     SollcScav 0    BatGosub m
WartDauer 1200  SolltEnd 3
SollcReakt 2.0    BatGosub m
SollcReakt 1.0    BatGosub m
Sollsweeps  50  SolltEnd 5
SollcReakt    0    BatGosub m
SollcReakt    0    Batgosub m
]
BatGosub w. [SP=2]
   SollcReakt 0
   WartDauer 5
   SollSweeps 50
   SolltEnd 2 [4096*0.488 ms]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:14, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
1 Traum     :  0.197     297.27 K
1 Tzelle    :  3.780     325.23 K     300 K
1 Tbad      :  0.002     272.27 K
1 TOHq      :  3.055     295.77 K    27.5125 mbar
1 TReakt    :  3.439     310.87 K     0.5237 mbar
1 pSat      :  7.470    1008.46 mbar
1 pZelle1000:  0.964     127.20 mbar   133 mbar
1 pZelle10  :  0.000       0.00 mbar
1 Flow1     :  2.585    1103.81 sccm
1 Flow2     :  2.940      32.78 sccm     Wasser:     1995.78     2000.00
1 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
1 Flow4     :  0.568      22.60 sccm        NO2:        0.50        0.50
1 Flow5     :  4.312     190.15 sccm
1       sumflow   :     1305.06 sccm  1300.00 sccm
1       dosierFlow:     1114.91 sccm  9999.00 sccm
1       flowZelle :       10393  ccm        0  ccm
1 4096*Dwell= 1.999 s    50 Sweeps
1 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/1]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:18, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
2 Traum     :  0.197     297.18 K
2 Tzelle    :  3.780     325.23 K     300 K
2 Tbad      :  0.001     272.10 K
2 TOHq      :  3.055     295.77 K    27.5182 mbar
2 TReakt    :  3.439     310.87 K     0.5235 mbar
2 pSat      :  7.470    1008.42 mbar
2 pZelle1000:  0.963     127.15 mbar   133 mbar
2 pZelle10  :  0.000       0.00 mbar
2 Flow1     :  3.014    1284.80 sccm
2 Flow2     :  3.022      33.66 sccm     Wasser:     2024.69     2000.00
2 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
2 Flow4     :  4.714     188.69 sccm        NO2:        0.00        0.00
2 Flow5     :  4.311     190.13 sccm
2       sumflow   :     1320.85 sccm  1300.00 sccm
2       dosierFlow:     1130.72 sccm  9999.00 sccm
2       flowZelle :       10523  ccm        0  ccm
2 4096*Dwell= 1.999 s    50 Sweeps
2 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/2]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:23, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
3 Traum     :  0.197     297.26 K
3 Tzelle    :  3.780     325.25 K     300 K
3 Tbad      :  0.001     272.18 K
3 TOHq      :  3.055     295.78 K    27.5346 mbar
3 TReakt    :  3.439     310.87 K     0.5236 mbar
3 pSat      :  7.469    1008.37 mbar
3 pZelle1000:  0.963     127.12 mbar   133 mbar
3 pZelle10  :  0.000       0.00 mbar
3 Flow1     :  2.974    1268.02 sccm
3 Flow2     :  2.941      32.78 sccm     Wasser:     2001.01     2000.00
3 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
3 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
3 Flow5     :  4.311     190.12 sccm
3       sumflow   :     1302.11 sccm  1300.00 sccm
3       dosierFlow:     1111.99 sccm  9999.00 sccm
3       flowZelle :       10376  ccm        0  ccm
3 4096*Dwell= 1.999 s    50 Sweeps
3 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/3]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:28, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
4 Traum     :  0.198     297.31 K
4 Tzelle    :  3.780     325.26 K     300 K
4 Tbad      :  0.001     272.19 K
4 TOHq      :  3.055     295.78 K    27.5372 mbar
4 TReakt    :  3.439     310.87 K     0.5237 mbar
4 pSat      :  7.469    1008.32 mbar
4 pZelle1000:  0.964     127.20 mbar   133 mbar
4 pZelle10  :  0.000       0.00 mbar
4 Flow1     :  2.974    1268.07 sccm
4 Flow2     :  2.939      32.76 sccm     Wasser:     2001.22     2000.00
4 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
4 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
4 Flow5     :  4.312     190.15 sccm
4       sumflow   :     1302.17 sccm  1300.00 sccm
4       dosierFlow:     1112.02 sccm  9999.00 sccm
4       flowZelle :       10370  ccm        0  ccm
4 4096*Dwell= 1.999 s    50 Sweeps
4 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/4]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:32, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
5 Traum     :  0.197     297.27 K
5 Tzelle    :  3.781     325.27 K     300 K
5 Tbad      :  0.001     272.19 K
5 TOHq      :  3.055     295.78 K    27.5359 mbar
5 TReakt    :  3.439     310.87 K     0.5236 mbar
5 pSat      :  7.469    1008.36 mbar
5 pZelle1000:  0.964     127.21 mbar   133 mbar
5 pZelle10  :  0.000       0.00 mbar
5 Flow1     :  2.974    1268.00 sccm
5 Flow2     :  2.940      32.77 sccm     Wasser:     2001.67     2000.00
5 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
5 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
5 Flow5     :  4.313     190.22 sccm
5       sumflow   :     1302.18 sccm  1300.00 sccm
5       dosierFlow:     1111.96 sccm  9999.00 sccm
5       flowZelle :       10369  ccm        0  ccm
5 4096*Dwell= 1.999 s    50 Sweeps
5 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/5]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:37, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
6 Traum     :  0.197     297.26 K
6 Tzelle    :  3.781     325.28 K     300 K
6 Tbad      :  0.001     272.18 K
6 TOHq      :  3.056     295.80 K    27.5637 mbar
6 TReakt    :  3.438     310.86 K     0.5232 mbar
6 pSat      :  7.469    1008.32 mbar
6 pZelle1000:  0.964     127.23 mbar   133 mbar
6 pZelle10  :  0.000       0.00 mbar
6 Flow1     :  2.974    1267.88 sccm
6 Flow2     :  2.940      32.77 sccm     Wasser:     2004.55     2000.00
6 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
6 Flow4     :  4.741     189.75 sccm        NO2:        0.00        0.00
6 Flow5     :  4.311     190.12 sccm
6       sumflow   :     1301.94 sccm  1300.00 sccm
6       dosierFlow:     1111.82 sccm  9999.00 sccm
6       flowZelle :       10366  ccm        0  ccm
6 4096*Dwell= 1.999 s    50 Sweeps
6 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/6]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:42, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
7 Traum     :  0.197     297.28 K
7 Tzelle    :  3.781     325.29 K     300 K
7 Tbad      :  0.001     272.18 K
7 TOHq      :  3.056     295.81 K    27.5732 mbar
7 TReakt    :  3.439     310.87 K     0.5237 mbar
7 pSat      :  7.470    1008.39 mbar
7 pZelle1000:  0.963     127.13 mbar   133 mbar
7 pZelle10  :  0.000       0.00 mbar
7 Flow1     :  2.974    1267.89 sccm
7 Flow2     :  2.941      32.79 sccm     Wasser:     2004.43     2000.00
7 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
7 Flow4     :  4.741     189.74 sccm        NO2:        0.00        0.00
7 Flow5     :  4.311     190.12 sccm
7       sumflow   :     1301.98 sccm  1300.00 sccm
7       dosierFlow:     1111.86 sccm  9999.00 sccm
7       flowZelle :       10374  ccm        0  ccm
7 4096*Dwell= 1.999 s    50 Sweeps
7 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/7]
      BatReturn  [SP=2]
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 16:46, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
8 Traum     :  0.198     297.31 K
8 Tzelle    :  3.782     325.31 K     300 K
8 Tbad      :  0.002     272.19 K
8 TOHq      :  3.055     295.80 K    27.5587 mbar
8 TReakt    :  3.439     310.87 K     0.5236 mbar
8 pSat      :  7.470    1008.44 mbar
8 pZelle1000:  0.963     127.17 mbar   133 mbar
8 pZelle10  :  0.000       0.00 mbar
8 Flow1     :  2.975    1268.22 sccm
8 Flow2     :  2.941      32.79 sccm     Wasser:     2003.47     2000.00
8 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
8 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
8 Flow5     :  4.310     190.08 sccm
8       sumflow   :     1302.29 sccm  1300.00 sccm
8       dosierFlow:     1112.20 sccm  9999.00 sccm
8       flowZelle :       10373  ccm        0  ccm
8 4096*Dwell= 1.999 s    50 Sweeps
8 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/8]
      BatReturn  [SP=2]
   WartDauer 1200
   SolltEnd 5 [4096*1.221 ms]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 17:11, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
9 Traum     :  0.196     297.04 K
9 Tzelle    :  3.783     325.34 K     300 K
9 Tbad      :  0.001     272.18 K
9 TOHq      :  3.055     295.77 K    27.5125 mbar
9 TReakt    :  3.438     310.86 K     0.5233 mbar
9 pSat      :  7.470    1008.50 mbar
9 pZelle1000:  0.964     127.26 mbar   133 mbar
9 pZelle10  :  0.000       0.00 mbar
9 Flow1     :  2.974    1267.97 sccm
9 Flow2     :  2.941      32.79 sccm     Wasser:     2000.97     2000.00
9 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
9 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
9 Flow5     :  4.312     190.15 sccm
9       sumflow   :     1302.11 sccm  1300.00 sccm
9       dosierFlow:     1111.96 sccm  9999.00 sccm
9       flowZelle :       10365  ccm        0  ccm
9 4096*Dwell= 5.001 s    50 Sweeps
9 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/9]
      BatReturn  [SP=2]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 24.8. 17:37, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
10 Traum     :  0.196     297.07 K
10 Tzelle    :  3.783     325.37 K     300 K
10 Tbad      :  0.002     272.19 K
10 TOHq      :  3.053     295.72 K    27.4253 mbar
10 TReakt    :  3.439     310.87 K     0.5237 mbar
10 pSat      :  7.469    1008.35 mbar
10 pZelle1000:  0.964     127.28 mbar   133 mbar
10 pZelle10  :  0.000       0.00 mbar
10 Flow1     :  2.974    1268.02 sccm
10 Flow2     :  2.940      32.78 sccm     Wasser:     1994.31     2000.00
10 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
10 Flow4     :  4.741     189.77 sccm        NO2:        0.00        0.00
10 Flow5     :  4.311     190.13 sccm
10       sumflow   :     1302.07 sccm  1300.00 sccm
10       dosierFlow:     1111.94 sccm  9999.00 sccm
10       flowZelle :       10363  ccm        0  ccm
10 4096*Dwell= 5.001 s    50 Sweeps
10 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/10]
      BatReturn  [SP=2]
   WartDauer 600
   SollSweeps 100
   SolltEnd 2 [4096*0.488 ms]
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 17:53, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
11 Traum     :  0.195     297.04 K
11 Tzelle    :  3.784     325.40 K     300 K
11 Tbad      :  0.002     272.21 K
11 TOHq      :  3.053     295.69 K    27.3725 mbar
11 TReakt    :  3.438     310.86 K     0.5234 mbar
11 pSat      :  7.470    1008.46 mbar
11 pZelle1000:  0.963     127.16 mbar   133 mbar
11 pZelle10  :  0.000       0.00 mbar
11 Flow1     :  2.936    1251.59 sccm
11 Flow2     :  2.940      32.77 sccm     Wasser:     1987.76     2000.00
11 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
11 Flow4     :  4.336     173.34 sccm        NO2:        0.05        0.05
11 Flow5     :  4.312     190.15 sccm
11       sumflow   :     1302.09 sccm  1300.00 sccm
11       dosierFlow:     1111.93 sccm  9999.00 sccm
11       flowZelle :       10373  ccm        0  ccm
11 4096*Dwell= 1.999 s    100 Sweeps
11 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/11]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 18:11, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
12 Traum     :  0.196     297.05 K
12 Tzelle    :  3.784     325.38 K     300 K
12 Tbad      :  0.002     272.24 K
12 TOHq      :  3.052     295.67 K    27.3510 mbar
12 TReakt    :  3.438     310.86 K     0.5234 mbar
12 pSat      :  7.470    1008.50 mbar
12 pZelle1000:  0.962     127.05 mbar   133 mbar
12 pZelle10  :  0.000       0.00 mbar
12 Flow1     :  2.897    1235.01 sccm
12 Flow2     :  2.981      33.22 sccm     Wasser:     2010.55     2000.00
12 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
12 Flow4     :  3.923     156.63 sccm        NO2:        0.10        0.10
12 Flow5     :  4.312     190.17 sccm
12       sumflow   :     1302.69 sccm  1300.00 sccm
12       dosierFlow:     1112.52 sccm  9999.00 sccm
12       flowZelle :       10387  ccm        0  ccm
12 4096*Dwell= 1.999 s    100 Sweeps
12 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/12]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 18:29, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
13 Traum     :  0.197     297.17 K
13 Tzelle    :  3.784     325.39 K     300 K
13 Tbad      :  0.002     272.26 K
13 TOHq      :  3.054     295.72 K    27.4367 mbar
13 TReakt    :  3.439     310.87 K     0.5238 mbar
13 pSat      :  7.470    1008.51 mbar
13 pZelle1000:  0.964     127.26 mbar   133 mbar
13 pZelle10  :  0.000       0.00 mbar
13 Flow1     :  2.700    1152.24 sccm
13 Flow2     :  2.941      32.79 sccm     Wasser:     2018.13     2000.00
13 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
13 Flow4     :  2.239      88.99 sccm        NO2:        0.30        0.30
13 Flow5     :  4.312     190.17 sccm
13       sumflow   :     1287.12 sccm  1300.00 sccm
13       dosierFlow:     1096.95 sccm  9999.00 sccm
13       flowZelle :       10246  ccm        0  ccm
13 4096*Dwell= 1.999 s    100 Sweeps
13 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/13]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 18:47, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
14 Traum     :  0.197     297.20 K
14 Tzelle    :  3.784     325.41 K     300 K
14 Tbad      :  0.002     272.27 K
14 TOHq      :  3.054     295.74 K    27.4664 mbar
14 TReakt    :  3.439     310.86 K     0.5235 mbar
14 pSat      :  7.471    1008.58 mbar
14 pZelle1000:  0.963     127.06 mbar   133 mbar
14 pZelle10  :  0.000       0.00 mbar
14 Flow1     :  2.544    1086.85 sccm
14 Flow2     :  2.941      32.79 sccm     Wasser:     2015.76     2000.00
14 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
14 Flow4     :  0.569      22.64 sccm        NO2:        0.50        0.50
14 Flow5     :  4.310     190.09 sccm
14       sumflow   :     1288.01 sccm  1300.00 sccm
14       dosierFlow:     1097.92 sccm  9999.00 sccm
14       flowZelle :       10269  ccm        0  ccm
14 4096*Dwell= 1.999 s    100 Sweeps
14 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/14]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 19:05, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
15 Traum     :  0.197     297.26 K
15 Tzelle    :  3.785     325.44 K     300 K
15 Tbad      :  0.002     272.28 K
15 TOHq      :  3.055     295.76 K    27.4948 mbar
15 TReakt    :  3.438     310.86 K     0.5233 mbar
15 pSat      :  7.472    1008.79 mbar
15 pZelle1000:  0.964     127.22 mbar   133 mbar
15 pZelle10  :  0.000       0.00 mbar
15 Flow1     :  2.544    1086.66 sccm
15 Flow2     :  2.940      32.78 sccm     Wasser:     2019.67     2000.00
15 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
15 Flow4     :  0.569      22.64 sccm        NO2:        0.50        0.50
15 Flow5     :  4.310     190.10 sccm
15       sumflow   :     1287.82 sccm  1300.00 sccm
15       dosierFlow:     1097.72 sccm  9999.00 sccm
15       flowZelle :       10254  ccm        0  ccm
15 4096*Dwell= 1.999 s    100 Sweeps
15 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/15]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 19:23, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
16 Traum     :  0.197     297.26 K
16 Tzelle    :  3.785     325.44 K     300 K
16 Tbad      :  0.002     272.26 K
16 TOHq      :  3.055     295.78 K    27.5372 mbar
16 TReakt    :  3.439     310.86 K     0.5235 mbar
16 pSat      :  7.475    1009.07 mbar
16 pZelle1000:  0.964     127.26 mbar   133 mbar
16 pZelle10  :  0.000       0.00 mbar
16 Flow1     :  2.701    1152.31 sccm
16 Flow2     :  2.940      32.77 sccm     Wasser:     2023.67     2000.00
16 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
16 Flow4     :  2.240      89.03 sccm        NO2:        0.30        0.30
16 Flow5     :  4.311     190.14 sccm
16       sumflow   :     1287.11 sccm  1300.00 sccm
16       dosierFlow:     1096.97 sccm  9999.00 sccm
16       flowZelle :       10245  ccm        0  ccm
16 4096*Dwell= 1.999 s    100 Sweeps
16 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/16]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 19:41, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
17 Traum     :  0.197     297.24 K
17 Tzelle    :  3.785     325.43 K     300 K
17 Tbad      :  0.002     272.24 K
17 TOHq      :  3.055     295.77 K    27.5137 mbar
17 TReakt    :  3.439     310.87 K     0.5235 mbar
17 pSat      :  7.476    1009.21 mbar
17 pZelle1000:  0.963     127.08 mbar   133 mbar
17 pZelle10  :  0.000       0.00 mbar
17 Flow1     :  2.897    1235.06 sccm
17 Flow2     :  2.943      32.81 sccm     Wasser:     1997.15     2000.00
17 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
17 Flow4     :  3.923     156.63 sccm        NO2:        0.10        0.10
17 Flow5     :  4.309     190.04 sccm
17       sumflow   :     1302.19 sccm  1300.00 sccm
17       dosierFlow:     1112.15 sccm  9999.00 sccm
17       flowZelle :       10381  ccm        0  ccm
17 4096*Dwell= 1.999 s    100 Sweeps
17 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/17]
      BatReturn  [SP=2]
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 19:59, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
18 Traum     :  0.197     297.16 K
18 Tzelle    :  3.785     325.45 K     300 K
18 Tbad      :  0.002     272.20 K
18 TOHq      :  3.055     295.77 K    27.5131 mbar
18 TReakt    :  3.438     310.86 K     0.5232 mbar
18 pSat      :  7.477    1009.41 mbar
18 pZelle1000:  0.964     127.21 mbar   133 mbar
18 pZelle10  :  0.000       0.00 mbar
18 Flow1     :  2.935    1251.51 sccm
18 Flow2     :  2.941      32.79 sccm     Wasser:     1997.95     2000.00
18 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
18 Flow4     :  4.337     173.36 sccm        NO2:        0.05        0.05
18 Flow5     :  4.311     190.10 sccm
18       sumflow   :     1301.95 sccm  1300.00 sccm
18       dosierFlow:     1111.85 sccm  9999.00 sccm
18       flowZelle :       10368  ccm        0  ccm
18 4096*Dwell= 1.999 s    100 Sweeps
18 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/18]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 20:16, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
19 Traum     :  0.196     297.07 K
19 Tzelle    :  3.784     325.42 K     300 K
19 Tbad      :  0.001     272.18 K
19 TOHq      :  3.054     295.75 K    27.4834 mbar
19 TReakt    :  3.439     310.87 K     0.5236 mbar
19 pSat      :  7.479    1009.68 mbar
19 pZelle1000:  0.963     127.18 mbar   133 mbar
19 pZelle10  :  0.000       0.00 mbar
19 Flow1     :  2.974    1268.02 sccm
19 Flow2     :  2.939      32.76 sccm     Wasser:     1993.24     2000.00
19 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
19 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
19 Flow5     :  4.311     190.10 sccm
19       sumflow   :     1302.08 sccm  1300.00 sccm
19       dosierFlow:     1111.97 sccm  9999.00 sccm
19       flowZelle :       10372  ccm        0  ccm
19 4096*Dwell= 1.999 s    100 Sweeps
19 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/19]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [100(100) Sweeps ab 24.8. 20:34, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
20 Traum     :  0.196     297.12 K
20 Tzelle    :  3.784     325.42 K     300 K
20 Tbad      :  0.002     272.19 K
20 TOHq      :  3.054     295.73 K    27.4468 mbar
20 TReakt    :  3.438     310.86 K     0.5232 mbar
20 pSat      :  7.480    1009.83 mbar
20 pZelle1000:  0.963     127.12 mbar   133 mbar
20 pZelle10  :  0.000       0.00 mbar
20 Flow1     :  2.974    1267.94 sccm
20 Flow2     :  2.942      32.79 sccm     Wasser:     1991.20     2000.00
20 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
20 Flow4     :  4.741     189.77 sccm        NO2:        0.00        0.00
20 Flow5     :  4.311     190.12 sccm
20       sumflow   :     1302.00 sccm  1300.00 sccm
20       dosierFlow:     1111.88 sccm  9999.00 sccm
20       flowZelle :       10376  ccm        0  ccm
20 4096*Dwell= 1.999 s    100 Sweeps
20 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/20]
      BatReturn  [SP=2]
   WartDauer 1200
   SollSweeps 200
   SolltEnd 3 [4096*0.732 ms]
   SollcReakt 1.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 21:02, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
21 Traum     :  0.197     297.22 K
21 Tzelle    :  3.785     325.45 K     300 K
21 Tbad      :  0.003     272.43 K
21 TOHq      :  3.054     295.75 K    27.4714 mbar
21 TReakt    :  3.439     310.90 K     0.5248 mbar
21 pSat      :  7.487    1010.70 mbar
21 pZelle1000:  0.969     127.92 mbar   133 mbar
21 pZelle10  :  0.000       0.00 mbar
21 Flow1     :  2.978    1269.53 sccm
21 Flow2     :  2.934      32.71 sccm     Wasser:     1994.87     2000.00
21 Flow3     :  0.678       0.91 sccm   m-Cresol:        1.03        1.00
21 Flow4     :  4.741     189.78 sccm        NO2:        0.00        0.00
21 Flow5     :  4.309     190.02 sccm
21       sumflow   :     1304.31 sccm  1300.00 sccm
21       dosierFlow:     1114.29 sccm  9999.00 sccm
21       flowZelle :       10329  ccm        0  ccm
21 4096*Dwell= 2.998 s    200 Sweeps
21 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/21]
      BatReturn  [SP=2]
   SollcReakt 2.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 21:37, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
22 Traum     :  0.197     297.28 K
22 Tzelle    :  3.785     325.45 K     300 K
22 Tbad      :  0.003     272.41 K
22 TOHq      :  3.054     295.73 K    27.4537 mbar
22 TReakt    :  3.439     310.89 K     0.5247 mbar
22 pSat      :  7.486    1010.67 mbar
22 pZelle1000:  0.969     127.91 mbar   133 mbar
22 pZelle10  :  0.000       0.00 mbar
22 Flow1     :  2.975    1268.14 sccm
22 Flow2     :  2.938      32.75 sccm     Wasser:     1996.71     2000.00
22 Flow3     :  1.449       1.77 sccm   m-Cresol:        2.01        2.00
22 Flow4     :  4.741     189.77 sccm        NO2:        0.00        0.00
22 Flow5     :  4.312     190.17 sccm
22       sumflow   :     1303.98 sccm  1300.00 sccm
22       dosierFlow:     1113.81 sccm  9999.00 sccm
22       flowZelle :       10327  ccm        0  ccm
22 4096*Dwell= 2.998 s    200 Sweeps
22 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/22]
      BatReturn  [SP=2]
   SolltEnd 2 [4096*0.488 ms]
   SollcReakt 3.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 22:12, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
23 Traum     :  0.197     297.20 K
23 Tzelle    :  3.785     325.43 K     300 K
23 Tbad      :  0.003     272.41 K
23 TOHq      :  3.054     295.72 K    27.4342 mbar
23 TReakt    :  3.439     310.89 K     0.5245 mbar
23 pSat      :  7.485    1010.53 mbar
23 pZelle1000:  0.969     127.88 mbar   133 mbar
23 pZelle10  :  0.000       0.00 mbar
23 Flow1     :  2.975    1268.21 sccm
23 Flow2     :  2.939      32.77 sccm     Wasser:     1994.75     2000.00
23 Flow3     :  2.307       2.69 sccm   m-Cresol:        3.04        3.00
23 Flow4     :  4.741     189.77 sccm        NO2:        0.00        0.00
23 Flow5     :  4.312     190.15 sccm
23       sumflow   :     1304.96 sccm  1300.00 sccm
23       dosierFlow:     1114.82 sccm  9999.00 sccm
23       flowZelle :       10338  ccm        0  ccm
23 4096*Dwell= 1.999 s    200 Sweeps
23 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/23]
      BatReturn  [SP=2]
   WartDauer 600
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 22:36, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
24 Traum     :  0.196     297.14 K
24 Tzelle    :  3.785     325.43 K     300 K
24 Tbad      :  0.003     272.43 K
24 TOHq      :  3.054     295.72 K    27.4323 mbar
24 TReakt    :  3.439     310.89 K     0.5246 mbar
24 pSat      :  7.486    1010.57 mbar
24 pZelle1000:  0.968     127.74 mbar   133 mbar
24 pZelle10  :  0.000       0.00 mbar
24 Flow1     :  2.935    1251.44 sccm
24 Flow2     :  2.939      32.76 sccm     Wasser:     1992.44     2000.00
24 Flow3     :  2.265       2.64 sccm   m-Cresol:        2.99        3.00
24 Flow4     :  4.336     173.33 sccm        NO2:        0.05        0.05
24 Flow5     :  4.313     190.19 sccm
24       sumflow   :     1304.63 sccm  1300.00 sccm
24       dosierFlow:     1114.44 sccm  9999.00 sccm
24       flowZelle :       10346  ccm        0  ccm
24 4096*Dwell= 1.999 s    200 Sweeps
24 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/24]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 23:01, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
25 Traum     :  0.198     297.32 K
25 Tzelle    :  3.785     325.43 K     300 K
25 Tbad      :  0.004     272.45 K
25 TOHq      :  3.054     295.72 K    27.4329 mbar
25 TReakt    :  3.439     310.89 K     0.5247 mbar
25 pSat      :  7.486    1010.60 mbar
25 pZelle1000:  0.969     127.92 mbar   133 mbar
25 pZelle10  :  0.000       0.00 mbar
25 Flow1     :  2.896    1234.96 sccm
25 Flow2     :  2.939      32.77 sccm     Wasser:     1995.09     2000.00
25 Flow3     :  2.265       2.64 sccm   m-Cresol:        2.99        3.00
25 Flow4     :  3.923     156.61 sccm        NO2:        0.10        0.10
25 Flow5     :  4.312     190.17 sccm
25       sumflow   :     1304.84 sccm  1300.00 sccm
25       dosierFlow:     1114.67 sccm  9999.00 sccm
25       flowZelle :       10333  ccm        0  ccm
25 4096*Dwell= 1.999 s    200 Sweeps
25 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/25]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 23:25, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
26 Traum     :  0.198     297.37 K
26 Tzelle    :  3.785     325.43 K     300 K
26 Tbad      :  0.004     272.52 K
26 TOHq      :  3.054     295.73 K    27.4500 mbar
26 TReakt    :  3.439     310.90 K     0.5247 mbar
26 pSat      :  7.485    1010.50 mbar
26 pZelle1000:  0.969     127.84 mbar   133 mbar
26 pZelle10  :  0.000       0.00 mbar
26 Flow1     :  2.739    1168.38 sccm
26 Flow2     :  2.939      32.76 sccm     Wasser:     1993.59     2000.00
26 Flow3     :  2.265       2.64 sccm   m-Cresol:        2.99        3.00
26 Flow4     :  2.239      88.97 sccm        NO2:        0.30        0.30
26 Flow5     :  4.311     190.12 sccm
26       sumflow   :     1305.85 sccm  1300.00 sccm
26       dosierFlow:     1115.73 sccm  9999.00 sccm
26       flowZelle :       10347  ccm        0  ccm
26 4096*Dwell= 1.999 s    200 Sweeps
26 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/26]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 24.8. 23:50, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
27 Traum     :  0.197     297.22 K
27 Tzelle    :  3.785     325.43 K     300 K
27 Tbad      :  0.004     272.50 K
27 TOHq      :  3.054     295.73 K    27.4487 mbar
27 TReakt    :  3.439     310.89 K     0.5246 mbar
27 pSat      :  7.485    1010.47 mbar
27 pZelle1000:  0.970     128.03 mbar   133 mbar
27 pZelle10  :  0.000       0.00 mbar
27 Flow1     :  2.546    1087.67 sccm
27 Flow2     :  2.940      32.78 sccm     Wasser:     2019.56     2000.00
27 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.03        3.00
27 Flow4     :  0.569      22.63 sccm        NO2:        0.50        0.50
27 Flow5     :  4.311     190.11 sccm
27       sumflow   :     1291.48 sccm  1300.00 sccm
27       dosierFlow:     1101.38 sccm  9999.00 sccm
27       flowZelle :       10218  ccm        0  ccm
27 4096*Dwell= 1.999 s    200 Sweeps
27 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/27]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 0:15, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
28 Traum     :  0.198     297.28 K
28 Tzelle    :  3.785     325.43 K     300 K
28 Tbad      :  0.004     272.50 K
28 TOHq      :  3.054     295.73 K    27.4405 mbar
28 TReakt    :  3.439     310.89 K     0.5245 mbar
28 pSat      :  7.485    1010.44 mbar
28 pZelle1000:  0.969     127.93 mbar   133 mbar
28 pZelle10  :  0.000       0.00 mbar
28 Flow1     :  2.546    1087.74 sccm
28 Flow2     :  2.939      32.77 sccm     Wasser:     2016.66     2000.00
28 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.03        3.00
28 Flow4     :  0.569      22.64 sccm        NO2:        0.50        0.50
28 Flow5     :  4.311     190.11 sccm
28       sumflow   :     1291.54 sccm  1300.00 sccm
28       dosierFlow:     1101.43 sccm  9999.00 sccm
28       flowZelle :       10227  ccm        0  ccm
28 4096*Dwell= 1.999 s    200 Sweeps
28 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/28]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 0:39, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
29 Traum     :  0.197     297.24 K
29 Tzelle    :  3.785     325.43 K     300 K
29 Tbad      :  0.004     272.50 K
29 TOHq      :  3.054     295.73 K    27.4468 mbar
29 TReakt    :  3.439     310.90 K     0.5250 mbar
29 pSat      :  7.487    1010.75 mbar
29 pZelle1000:  0.970     128.03 mbar   133 mbar
29 pZelle10  :  0.000       0.00 mbar
29 Flow1     :  2.739    1168.32 sccm
29 Flow2     :  2.939      32.77 sccm     Wasser:     1996.11     2000.00
29 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.00        3.00
29 Flow4     :  2.238      88.96 sccm        NO2:        0.30        0.30
29 Flow5     :  4.312     190.15 sccm
29       sumflow   :     1305.83 sccm  1300.00 sccm
29       dosierFlow:     1115.69 sccm  9999.00 sccm
29       flowZelle :       10332  ccm        0  ccm
29 4096*Dwell= 1.999 s    200 Sweeps
29 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/29]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 1:04, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
30 Traum     :  0.198     297.29 K
30 Tzelle    :  3.785     325.44 K     300 K
30 Tbad      :  0.004     272.45 K
30 TOHq      :  3.054     295.73 K    27.4474 mbar
30 TReakt    :  3.439     310.89 K     0.5244 mbar
30 pSat      :  7.483    1010.27 mbar
30 pZelle1000:  0.969     127.90 mbar   133 mbar
30 pZelle10  :  0.000       0.00 mbar
30 Flow1     :  2.896    1234.71 sccm
30 Flow2     :  2.939      32.77 sccm     Wasser:     1997.25     2000.00
30 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.00        3.00
30 Flow4     :  3.923     156.62 sccm        NO2:        0.10        0.10
30 Flow5     :  4.310     190.09 sccm
30       sumflow   :     1304.51 sccm  1300.00 sccm
30       dosierFlow:     1114.42 sccm  9999.00 sccm
30       flowZelle :       10332  ccm        0  ccm
30 4096*Dwell= 1.999 s    200 Sweeps
30 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/30]
      BatReturn  [SP=2]
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 1:28, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
31 Traum     :  0.197     297.20 K
31 Tzelle    :  3.784     325.42 K     300 K
31 Tbad      :  0.004     272.45 K
31 TOHq      :  3.054     295.73 K    27.4481 mbar
31 TReakt    :  3.439     310.90 K     0.5247 mbar
31 pSat      :  7.482    1010.13 mbar
31 pZelle1000:  0.969     127.89 mbar   133 mbar
31 pZelle10  :  0.000       0.00 mbar
31 Flow1     :  2.935    1251.48 sccm
31 Flow2     :  2.939      32.77 sccm     Wasser:     1997.43     2000.00
31 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.00        3.00
31 Flow4     :  4.336     173.36 sccm        NO2:        0.05        0.05
31 Flow5     :  4.311     190.10 sccm
31       sumflow   :     1304.55 sccm  1300.00 sccm
31       dosierFlow:     1114.45 sccm  9999.00 sccm
31       flowZelle :       10333  ccm        0  ccm
31 4096*Dwell= 1.999 s    200 Sweeps
31 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/31]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 1:53, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
32 Traum     :  0.198     297.33 K
32 Tzelle    :  3.785     325.46 K     300 K
32 Tbad      :  0.003     272.40 K
32 TOHq      :  3.054     295.74 K    27.4582 mbar
32 TReakt    :  3.440     310.90 K     0.5251 mbar
32 pSat      :  7.480    1009.75 mbar
32 pZelle1000:  0.969     127.96 mbar   133 mbar
32 pZelle10  :  0.000       0.00 mbar
32 Flow1     :  2.974    1268.02 sccm
32 Flow2     :  2.939      32.76 sccm     Wasser:     1998.95     2000.00
32 Flow3     :  2.304       2.68 sccm   m-Cresol:        3.05        3.00
32 Flow4     :  4.740     189.72 sccm        NO2:        0.00        0.00
32 Flow5     :  4.312     190.17 sccm
32       sumflow   :     1304.84 sccm  1300.00 sccm
32       dosierFlow:     1114.67 sccm  9999.00 sccm
32       flowZelle :       10330  ccm        0  ccm
32 4096*Dwell= 1.999 s    200 Sweeps
32 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/32]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 2:18, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
33 Traum     :  0.198     297.33 K
33 Tzelle    :  3.784     325.42 K     300 K
33 Tbad      :  0.003     272.41 K
33 TOHq      :  3.054     295.73 K    27.4519 mbar
33 TReakt    :  3.439     310.89 K     0.5245 mbar
33 pSat      :  7.478    1009.48 mbar
33 pZelle1000:  0.969     127.90 mbar   133 mbar
33 pZelle10  :  0.000       0.00 mbar
33 Flow1     :  2.975    1268.10 sccm
33 Flow2     :  2.939      32.77 sccm     Wasser:     1998.65     2000.00
33 Flow3     :  2.264       2.64 sccm   m-Cresol:        3.00        3.00
33 Flow4     :  4.741     189.76 sccm        NO2:        0.00        0.00
33 Flow5     :  4.312     190.18 sccm
33       sumflow   :     1304.84 sccm  1300.00 sccm
33       dosierFlow:     1114.67 sccm  9999.00 sccm
33       flowZelle :       10334  ccm        0  ccm
33 4096*Dwell= 1.999 s    200 Sweeps
33 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/33]
      BatReturn  [SP=2]
   WartDauer 1200
   SollcReakt 4.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 2:52, 1 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
34 Traum     :  0.198     297.39 K
34 Tzelle    :  3.786     325.49 K     300 K
34 Tbad      :  0.003     272.40 K
34 TOHq      :  3.054     295.74 K    27.4576 mbar
34 TReakt    :  3.439     310.89 K     0.5245 mbar
34 pSat      :  7.474    1009.05 mbar
34 pZelle1000:  0.969     127.84 mbar   133 mbar
34 pZelle10  :  0.000       0.00 mbar
34 Flow1     :  2.975    1268.42 sccm
34 Flow2     :  2.941      32.79 sccm     Wasser:     1998.18     2000.00
34 Flow3     :  3.164       3.56 sccm   m-Cresol:        4.03        4.00
34 Flow4     :  4.741     189.75 sccm        NO2:        0.00        0.00
34 Flow5     :  4.311     190.12 sccm
34       sumflow   :     1306.06 sccm  1300.00 sccm
34       dosierFlow:     1115.94 sccm  9999.00 sccm
34       flowZelle :       10349  ccm        0  ccm
34 4096*Dwell= 1.999 s    200 Sweeps
34 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/34]
      BatReturn  [SP=2]
   SollSweeps 300
   SolltEnd 1 [4096*0.244 ms]
   SollcReakt 5.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [300(300) Sweeps ab 25.8. 3:27, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
35 Traum     :  0.198     297.29 K
35 Tzelle    :  3.785     325.44 K     300 K
35 Tbad      :  0.003     272.41 K
35 TOHq      :  3.054     295.74 K    27.4556 mbar
35 TReakt    :  3.439     310.90 K     0.5249 mbar
35 pSat      :  7.471    1008.65 mbar
35 pZelle1000:  0.969     127.87 mbar   133 mbar
35 pZelle10  :  0.000       0.00 mbar
35 Flow1     :  2.976    1268.53 sccm
35 Flow2     :  2.942      32.80 sccm     Wasser:     1998.52     2000.00
35 Flow3     :  4.063       4.42 sccm   m-Cresol:        5.01        5.00
35 Flow4     :  4.741     189.75 sccm        NO2:        0.00        0.00
35 Flow5     :  4.311     190.10 sccm
35       sumflow   :     1307.01 sccm  1300.00 sccm
35       dosierFlow:     1116.91 sccm  9999.00 sccm
35       flowZelle :       10355  ccm        0  ccm
35 4096*Dwell= 0.999 s    300 Sweeps
35 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/35]
      BatReturn  [SP=2]
   SollcReakt 6.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [300(300) Sweeps ab 25.8. 4:08, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
36 Traum     :  0.198     297.30 K
36 Tzelle    :  3.785     325.43 K     300 K
36 Tbad      :  0.003     272.40 K
36 TOHq      :  3.054     295.74 K    27.4576 mbar
36 TReakt    :  3.439     310.90 K     0.5249 mbar
36 pSat      :  7.469    1008.34 mbar
36 pZelle1000:  0.968     127.83 mbar   133 mbar
36 pZelle10  :  0.000       0.00 mbar
36 Flow1     :  2.976    1268.57 sccm
36 Flow2     :  2.941      32.79 sccm     Wasser:     1996.36     2000.00
36 Flow3     :  5.004       5.27 sccm   m-Cresol:        5.97        6.00
36 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
36 Flow5     :  4.313     190.19 sccm
36       sumflow   :     1308.01 sccm  1300.00 sccm
36       dosierFlow:     1117.81 sccm  9999.00 sccm
36       flowZelle :       10366  ccm        0  ccm
36 4096*Dwell= 0.999 s    300 Sweeps
36 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/36]
      BatReturn  [SP=2]
   SollcReakt 6.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [300(300) Sweeps ab 25.8. 4:50, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
37 Traum     :  0.198     297.32 K
37 Tzelle    :  3.785     325.44 K     300 K
37 Tbad      :  0.003     272.39 K
37 TOHq      :  3.054     295.75 K    27.4841 mbar
37 TReakt    :  3.439     310.89 K     0.5246 mbar
37 pSat      :  7.464    1007.68 mbar
37 pZelle1000:  0.968     127.81 mbar   133 mbar
37 pZelle10  :  0.000       0.00 mbar
37 Flow1     :  2.975    1268.33 sccm
37 Flow2     :  2.940      32.78 sccm     Wasser:     1999.55     2000.00
37 Flow3     :  5.002       5.26 sccm   m-Cresol:        5.97        6.00
37 Flow4     :  4.741     189.75 sccm        NO2:        0.00        0.00
37 Flow5     :  4.311     190.11 sccm
37       sumflow   :     1307.65 sccm  1300.00 sccm
37       dosierFlow:     1117.55 sccm  9999.00 sccm
37       flowZelle :       10364  ccm        0  ccm
37 4096*Dwell= 0.999 s    300 Sweeps
37 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/37]
      BatReturn  [SP=2]
   SollcReakt 5.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [300(300) Sweeps ab 25.8. 5:31, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
38 Traum     :  0.198     297.39 K
38 Tzelle    :  3.786     325.48 K     300 K
38 Tbad      :  0.003     272.40 K
38 TOHq      :  3.054     295.76 K    27.4897 mbar
38 TReakt    :  3.439     310.90 K     0.5249 mbar
38 pSat      :  7.461    1007.22 mbar
38 pZelle1000:  0.968     127.80 mbar   133 mbar
38 pZelle10  :  0.000       0.00 mbar
38 Flow1     :  2.975    1268.47 sccm
38 Flow2     :  2.941      32.79 sccm     Wasser:     2002.01     2000.00
38 Flow3     :  4.064       4.42 sccm   m-Cresol:        5.01        5.00
38 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
38 Flow5     :  4.310     190.08 sccm
38       sumflow   :     1306.94 sccm  1300.00 sccm
38       dosierFlow:     1116.86 sccm  9999.00 sccm
38       flowZelle :       10359  ccm        0  ccm
38 4096*Dwell= 0.999 s    300 Sweeps
38 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/38]
      BatReturn  [SP=2]
   SollSweeps 200
   SolltEnd 2 [4096*0.488 ms]
   SollcReakt 4.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 6:12, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
39 Traum     :  0.198     297.34 K
39 Tzelle    :  3.785     325.42 K     300 K
39 Tbad      :  0.003     272.40 K
39 TOHq      :  3.055     295.77 K    27.5201 mbar
39 TReakt    :  3.439     310.90 K     0.5249 mbar
39 pSat      :  7.459    1006.94 mbar
39 pZelle1000:  0.968     127.77 mbar   133 mbar
39 pZelle10  :  0.000       0.00 mbar
39 Flow1     :  2.975    1268.36 sccm
39 Flow2     :  2.940      32.78 sccm     Wasser:     2005.40     2000.00
39 Flow3     :  3.165       3.56 sccm   m-Cresol:        4.04        4.00
39 Flow4     :  4.740     189.71 sccm        NO2:        0.00        0.00
39 Flow5     :  4.311     190.11 sccm
39       sumflow   :     1306.03 sccm  1300.00 sccm
39       dosierFlow:     1115.91 sccm  9999.00 sccm
39       flowZelle :       10355  ccm        0  ccm
39 4096*Dwell= 1.999 s    200 Sweeps
39 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/39]
      BatReturn  [SP=2]
   SollcReakt 3.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 6:47, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
40 Traum     :  0.199     297.51 K
40 Tzelle    :  3.785     325.44 K     300 K
40 Tbad      :  0.003     272.43 K
40 TOHq      :  3.055     295.78 K    27.5238 mbar
40 TReakt    :  3.439     310.89 K     0.5245 mbar
40 pSat      :  7.457    1006.72 mbar
40 pZelle1000:  0.969     127.84 mbar   133 mbar
40 pZelle10  :  0.000       0.00 mbar
40 Flow1     :  2.975    1268.22 sccm
40 Flow2     :  2.939      32.77 sccm     Wasser:     2008.71     2000.00
40 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.00        3.00
40 Flow4     :  4.741     189.78 sccm        NO2:        0.00        0.00
40 Flow5     :  4.311     190.12 sccm
40       sumflow   :     1304.89 sccm  1300.00 sccm
40       dosierFlow:     1114.77 sccm  9999.00 sccm
40       flowZelle :       10340  ccm        0  ccm
40 4096*Dwell= 1.999 s    200 Sweeps
40 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/40]
      BatReturn  [SP=2]
   WartDauer 600
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 7:11, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
41 Traum     :  0.199     297.42 K
41 Tzelle    :  3.785     325.44 K     300 K
41 Tbad      :  0.004     272.45 K
41 TOHq      :  3.055     295.78 K    27.5315 mbar
41 TReakt    :  3.439     310.89 K     0.5245 mbar
41 pSat      :  7.456    1006.49 mbar
41 pZelle1000:  0.969     127.90 mbar   133 mbar
41 pZelle10  :  0.000       0.00 mbar
41 Flow1     :  2.936    1251.78 sccm
41 Flow2     :  2.940      32.78 sccm     Wasser:     2011.17     2000.00
41 Flow3     :  2.264       2.64 sccm   m-Cresol:        3.00        3.00
41 Flow4     :  4.337     173.40 sccm        NO2:        0.05        0.05
41 Flow5     :  4.311     190.12 sccm
41       sumflow   :     1304.85 sccm  1300.00 sccm
41       dosierFlow:     1114.72 sccm  9999.00 sccm
41       flowZelle :       10335  ccm        0  ccm
41 4096*Dwell= 1.999 s    200 Sweeps
41 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/41]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 7:36, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
42 Traum     :  0.198     297.38 K
42 Tzelle    :  3.785     325.43 K     300 K
42 Tbad      :  0.004     272.45 K
42 TOHq      :  3.056     295.80 K    27.5701 mbar
42 TReakt    :  3.439     310.90 K     0.5247 mbar
42 pSat      :  7.455    1006.43 mbar
42 pZelle1000:  0.969     127.89 mbar   133 mbar
42 pZelle10  :  0.000       0.00 mbar
42 Flow1     :  2.896    1234.55 sccm
42 Flow2     :  2.939      32.76 sccm     Wasser:     2014.07     2000.00
42 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.01        3.00
42 Flow4     :  3.923     156.62 sccm        NO2:        0.10        0.10
42 Flow5     :  4.310     190.08 sccm
42       sumflow   :     1304.34 sccm  1300.00 sccm
42       dosierFlow:     1114.26 sccm  9999.00 sccm
42       flowZelle :       10332  ccm        0  ccm
42 4096*Dwell= 1.999 s    200 Sweeps
42 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/42]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 8:00, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
43 Traum     :  0.199     297.41 K
43 Tzelle    :  3.784     325.42 K     300 K
43 Tbad      :  0.004     272.51 K
43 TOHq      :  3.056     295.81 K    27.5770 mbar
43 TReakt    :  3.439     310.90 K     0.5249 mbar
43 pSat      :  7.454    1006.29 mbar
43 pZelle1000:  0.969     127.85 mbar   133 mbar
43 pZelle10  :  0.000       0.00 mbar
43 Flow1     :  2.739    1168.38 sccm
43 Flow2     :  2.897      32.32 sccm     Wasser:     1985.01     2000.00
43 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.01        3.00
43 Flow4     :  2.238      88.95 sccm        NO2:        0.30        0.30
43 Flow5     :  4.312     190.17 sccm
43       sumflow   :     1305.48 sccm  1300.00 sccm
43       dosierFlow:     1115.30 sccm  9999.00 sccm
43       flowZelle :       10344  ccm        0  ccm
43 4096*Dwell= 1.999 s    200 Sweeps
43 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/43]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 8:25, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
44 Traum     :  0.199     297.52 K
44 Tzelle    :  3.785     325.43 K     300 K
44 Tbad      :  0.004     272.50 K
44 TOHq      :  3.056     295.82 K    27.5916 mbar
44 TReakt    :  3.439     310.89 K     0.5246 mbar
44 pSat      :  7.453    1006.16 mbar
44 pZelle1000:  0.969     127.96 mbar   133 mbar
44 pZelle10  :  0.000       0.00 mbar
44 Flow1     :  2.545    1087.14 sccm
44 Flow2     :  2.896      32.31 sccm     Wasser:     2010.44     2000.00
44 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.04        3.00
44 Flow4     :  0.570      22.66 sccm        NO2:        0.50        0.50
44 Flow5     :  4.312     190.17 sccm
44       sumflow   :     1290.51 sccm  1300.00 sccm
44       dosierFlow:     1100.34 sccm  9999.00 sccm
44       flowZelle :       10217  ccm        0  ccm
44 4096*Dwell= 1.999 s    200 Sweeps
44 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/44]
      BatReturn  [SP=2]
   SollcScav 0.5
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 8:50, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
45 Traum     :  0.199     297.42 K
45 Tzelle    :  3.784     325.42 K     300 K
45 Tbad      :  0.004     272.50 K
45 TOHq      :  3.056     295.82 K    27.5954 mbar
45 TReakt    :  3.439     310.89 K     0.5245 mbar
45 pSat      :  7.454    1006.23 mbar
45 pZelle1000:  0.970     128.00 mbar   133 mbar
45 pZelle10  :  0.000       0.00 mbar
45 Flow1     :  2.545    1087.15 sccm
45 Flow2     :  2.897      32.32 sccm     Wasser:     2011.91     2000.00
45 Flow3     :  2.263       2.64 sccm   m-Cresol:        3.04        3.00
45 Flow4     :  0.570      22.65 sccm        NO2:        0.50        0.50
45 Flow5     :  4.310     190.09 sccm
45       sumflow   :     1290.46 sccm  1300.00 sccm
45       dosierFlow:     1100.37 sccm  9999.00 sccm
45       flowZelle :       10213  ccm        0  ccm
45 4096*Dwell= 1.999 s    200 Sweeps
45 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/45]
      BatReturn  [SP=2]
   SollcScav 0.3
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 9:14, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
46 Traum     :  0.198     297.36 K
46 Tzelle    :  3.785     325.44 K     300 K
46 Tbad      :  0.004     272.49 K
46 TOHq      :  3.056     295.82 K    27.5942 mbar
46 TReakt    :  3.439     310.90 K     0.5247 mbar
46 pSat      :  7.454    1006.28 mbar
46 pZelle1000:  0.970     128.02 mbar   133 mbar
46 pZelle10  :  0.000       0.00 mbar
46 Flow1     :  2.740    1168.72 sccm
46 Flow2     :  2.897      32.31 sccm     Wasser:     1988.14     2000.00
46 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.01        3.00
46 Flow4     :  2.240      89.02 sccm        NO2:        0.30        0.30
46 Flow5     :  4.312     190.15 sccm
46       sumflow   :     1305.71 sccm  1300.00 sccm
46       dosierFlow:     1115.56 sccm  9999.00 sccm
46       flowZelle :       10332  ccm        0  ccm
46 4096*Dwell= 1.999 s    200 Sweeps
46 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/46]
      BatReturn  [SP=2]
   SollcScav 0.1
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 9:39, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
47 Traum     :  0.198     297.32 K
47 Tzelle    :  3.784     325.42 K     300 K
47 Tbad      :  0.004     272.44 K
47 TOHq      :  3.056     295.83 K    27.6183 mbar
47 TReakt    :  3.439     310.89 K     0.5246 mbar
47 pSat      :  7.453    1006.12 mbar
47 pZelle1000:  0.969     127.88 mbar   133 mbar
47 pZelle10  :  0.000       0.00 mbar
47 Flow1     :  2.896    1234.92 sccm
47 Flow2     :  2.939      32.77 sccm     Wasser:     2017.88     2000.00
47 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.01        3.00
47 Flow4     :  3.923     156.63 sccm        NO2:        0.10        0.10
47 Flow5     :  4.311     190.11 sccm
47       sumflow   :     1304.74 sccm  1300.00 sccm
47       dosierFlow:     1114.62 sccm  9999.00 sccm
47       flowZelle :       10335  ccm        0  ccm
47 4096*Dwell= 1.999 s    200 Sweeps
47 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/47]
      BatReturn  [SP=2]
   SollcScav 0.05
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 10:04, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
48 Traum     :  0.198     297.33 K
48 Tzelle    :  3.785     325.42 K     300 K
48 Tbad      :  0.003     272.43 K
48 TOHq      :  3.057     295.86 K    27.6684 mbar
48 TReakt    :  3.439     310.89 K     0.5247 mbar
48 pSat      :  7.454    1006.27 mbar
48 pZelle1000:  0.969     127.93 mbar   133 mbar
48 pZelle10  :  0.000       0.00 mbar
48 Flow1     :  2.935    1251.38 sccm
48 Flow2     :  2.898      32.32 sccm     Wasser:     1995.51     2000.00
48 Flow3     :  2.266       2.64 sccm   m-Cresol:        3.01        3.00
48 Flow4     :  4.338     173.40 sccm        NO2:        0.05        0.05
48 Flow5     :  4.313     190.20 sccm
48       sumflow   :     1304.05 sccm  1300.00 sccm
48       dosierFlow:     1113.86 sccm  9999.00 sccm
48       flowZelle :       10326  ccm        0  ccm
48 4096*Dwell= 1.999 s    200 Sweeps
48 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/48]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 10:28, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
49 Traum     :  0.198     297.38 K
49 Tzelle    :  3.784     325.41 K     300 K
49 Tbad      :  0.003     272.42 K
49 TOHq      :  3.059     295.92 K    27.7607 mbar
49 TReakt    :  3.439     310.90 K     0.5248 mbar
49 pSat      :  7.454    1006.24 mbar
49 pZelle1000:  0.968     127.84 mbar   133 mbar
49 pZelle10  :  0.000       0.00 mbar
49 Flow1     :  2.975    1268.22 sccm
49 Flow2     :  2.897      32.31 sccm     Wasser:     1999.88     2000.00
49 Flow3     :  2.264       2.64 sccm   m-Cresol:        3.01        3.00
49 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
49 Flow5     :  4.313     190.20 sccm
49       sumflow   :     1304.56 sccm  1300.00 sccm
49       dosierFlow:     1114.35 sccm  9999.00 sccm
49       flowZelle :       10337  ccm        0  ccm
49 4096*Dwell= 1.999 s    200 Sweeps
49 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/49]
      BatReturn  [SP=2]
   SollcScav 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 10:53, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
50 Traum     :  0.199     297.42 K
50 Tzelle    :  3.784     325.39 K     300 K
50 Tbad      :  0.003     272.42 K
50 TOHq      :  3.059     295.94 K    27.7932 mbar
50 TReakt    :  3.439     310.90 K     0.5249 mbar
50 pSat      :  7.450    1005.79 mbar
50 pZelle1000:  0.969     127.94 mbar   133 mbar
50 pZelle10  :  0.000       0.00 mbar
50 Flow1     :  2.975    1268.22 sccm
50 Flow2     :  2.897      32.31 sccm     Wasser:     2004.78     2000.00
50 Flow3     :  2.265       2.64 sccm   m-Cresol:        3.01        3.00
50 Flow4     :  4.740     189.73 sccm        NO2:        0.00        0.00
50 Flow5     :  4.314     190.25 sccm
50       sumflow   :     1304.61 sccm  1300.00 sccm
50       dosierFlow:     1114.36 sccm  9999.00 sccm
50       flowZelle :       10330  ccm        0  ccm
50 4096*Dwell= 1.999 s    200 Sweeps
50 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/50]
      BatReturn  [SP=2]
   WartDauer 1200
   SolltEnd 3 [4096*0.732 ms]
   SollcReakt 2.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 11:27, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
51 Traum     :  0.199     297.48 K
51 Tzelle    :  3.784     325.41 K     300 K
51 Tbad      :  0.003     272.43 K
51 TOHq      :  3.060     295.98 K    27.8666 mbar
51 TReakt    :  3.439     310.90 K     0.5248 mbar
51 pSat      :  7.451    1005.87 mbar
51 pZelle1000:  0.969     127.94 mbar   133 mbar
51 pZelle10  :  0.000       0.00 mbar
51 Flow1     :  2.974    1267.81 sccm
51 Flow2     :  2.895      32.29 sccm     Wasser:     2010.79     2000.00
51 Flow3     :  1.449       1.77 sccm   m-Cresol:        2.02        2.00
51 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
51 Flow5     :  4.311     190.12 sccm
51       sumflow   :     1303.18 sccm  1300.00 sccm
51       dosierFlow:     1113.05 sccm  9999.00 sccm
51       flowZelle :       10318  ccm        0  ccm
51 4096*Dwell= 2.998 s    200 Sweeps
51 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/51]
      BatReturn  [SP=2]
   SollcReakt 1.0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [200(200) Sweeps ab 25.8. 12:02, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
52 Traum     :  0.200     297.57 K
52 Tzelle    :  3.784     325.42 K     300 K
52 Tbad      :  0.003     272.43 K
52 TOHq      :  3.060     295.99 K    27.8897 mbar
52 TReakt    :  3.439     310.90 K     0.5247 mbar
52 pSat      :  7.450    1005.75 mbar
52 pZelle1000:  0.969     127.95 mbar   133 mbar
52 pZelle10  :  0.000       0.00 mbar
52 Flow1     :  2.977    1268.97 sccm
52 Flow2     :  2.891      32.25 sccm     Wasser:     2009.97     2000.00
52 Flow3     :  0.678       0.91 sccm   m-Cresol:        1.03        1.00
52 Flow4     :  4.740     189.74 sccm        NO2:        0.00        0.00
52 Flow5     :  4.312     190.15 sccm
52       sumflow   :     1303.46 sccm  1300.00 sccm
52       dosierFlow:     1113.32 sccm  9999.00 sccm
52       flowZelle :       10320  ccm        0  ccm
52 4096*Dwell= 2.998 s    200 Sweeps
52 BlitzFolge= 4.000 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/52]
      BatReturn  [SP=2]
   SollSweeps 50
   SolltEnd 5 [4096*1.221 ms]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 25.8. 12:40, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
53 Traum     :  0.198     297.33 K
53 Tzelle    :  3.785     325.43 K     300 K
53 Tbad      :  0.002     272.20 K
53 TOHq      :  3.060     295.98 K    27.8641 mbar
53 TReakt    :  3.438     310.86 K     0.5232 mbar
53 pSat      :  7.451    1005.84 mbar
53 pZelle1000:  0.964     127.28 mbar   133 mbar
53 pZelle10  :  0.000       0.00 mbar
53 Flow1     :  2.975    1268.22 sccm
53 Flow2     :  2.900      32.35 sccm     Wasser:     2005.67     2000.00
53 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
53 Flow4     :  4.740     189.72 sccm        NO2:        0.00        0.00
53 Flow5     :  4.310     190.07 sccm
53       sumflow   :     1301.84 sccm  1300.00 sccm
53       dosierFlow:     1111.77 sccm  9999.00 sccm
53       flowZelle :       10361  ccm        0  ccm
53 4096*Dwell= 5.001 s    50 Sweeps
53 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/53]
      BatReturn  [SP=2]
   SollcReakt 0
   BatGosub m [SP=3]
      EinstKonz 
      WartAnfang 
      WartWarte 
      EinstKonz 
      Messung  [50(50) Sweeps ab 25.8. 13:06, 0 Fehltrigger]
      N2off 
      ProtIstWerte [
             Spannung    Istwert
54 Traum     :  0.198     297.32 K
54 Tzelle    :  3.783     325.38 K     300 K
54 Tbad      :  0.001     272.18 K
54 TOHq      :  3.060     295.98 K    27.8596 mbar
54 TReakt    :  3.438     310.86 K     0.5233 mbar
54 pSat      :  7.452    1005.98 mbar
54 pZelle1000:  0.963     127.16 mbar   133 mbar
54 pZelle10  :  0.000       0.00 mbar
54 Flow1     :  2.975    1268.36 sccm
54 Flow2     :  2.899      32.33 sccm     Wasser:     2002.47     2000.00
54 Flow3     : -0.019       0.00 sccm   m-Cresol:        0.00        0.00
54 Flow4     :  4.741     189.76 sccm        NO2:        0.00        0.00
54 Flow5     :  4.309     190.05 sccm
54       sumflow   :     1301.91 sccm  1300.00 sccm
54       dosierFlow:     1111.86 sccm  9999.00 sccm
54       flowZelle :       10372  ccm        0  ccm
54 4096*Dwell= 5.001 s    50 Sweeps
54 BlitzFolge= 5.001 s    BlitzEnergie  2.40 J ]
  [Fehler: ADC-Overflow (>9.5V)]
      SaveMeas  [7/54]
      BatReturn  [SP=2]
   BatReturn  [SP=1]
DosBefehl :
